text:
Electrocardiography
From Wikipedia, the free encyclopedia
Jump to navigation Jump to search
"ECG" and "EKG" redirect here. For other uses, see ECG (disambiguation) and EKG (disambiguation).
Not to be confused with other types of electrography or with echocardiography.
Electrocardiography
ECG of a heart in normal sinus rhythm
ICD-10-PCS
R94.31
ICD-9-CM
89.52
MeSH
D004562
MedlinePlus
003868
[edit on Wikidata]
Electrocardiography (ECG or EKG[a]) is the process of recording the electrical activity of the heart[4] over a period of time using electrodes placed over the skin. These electrodes detect the tiny electrical changes on the skin that arise from the heart muscle's electrophysiologic pattern of depolarizing and repolarizing during each heartbeat. It is very commonly performed to detect any cardiac problems.
In a conventional 12-lead ECG, ten electrodes are placed on the patient's limbs and on the surface of the chest. The overall magnitude of the heart's electrical potential is then measured from twelve different angles ("leads") and is recorded over a period of time (usually ten seconds). In this way, the overall magnitude and direction of the heart's electrical depolarization is captured at each moment throughout the cardiac cycle.[5] The graph of voltage versus time produced by this noninvasive medical procedure is an electrocardiogram.
There are three main components to an ECG: the P wave, which represents the depolarization of the atria; the QRS complex, which represents the depolarization of the ventricles; and the T wave, which represents the repolarization of the ventricles.[6] It can also be further broken down into the following:
O is the origin or datum point preceding the cycle
P is the atrial systole contraction pulse
Q is a downward deflection immediately preceding the ventricular contraction
R is the peak of the ventricular contraction
S is the downward deflection immediately after the ventricular contraction
T is the recovery of the ventricles
U is the successor of the T wave but it is small and not always observed
During each heartbeat, a healthy heart has an orderly progression of depolarization that starts with pacemaker cells in the sinoatrial node, spreads throughout the atrium, passes through the atrioventricular node down into the bundle of His and into the Purkinje fibers, spreading down and to the left throughout the ventricles.[6] This orderly pattern of depolarization gives rise to the characteristic ECG tracing. To the trained clinician, an ECG conveys a large amount of information about the structure of the heart and the function of its electrical conduction system.[7] Among other things, an ECG can be used to measure the rate and rhythm of heartbeats, the size and position of the heart chambers, the presence of any damage to the heart's muscle cells or conduction system, the effects of heart drugs, and the function of implanted pacemakers.[8]
Contents
1 Medical uses
2 Electrocardiographs
3 Electrodes and leads
3.1 Limb leads
3.2 Augmented limb leads
3.3 Precordial leads
3.4 Specialized leads
3.5 Lead locations on an ECG report
3.6 Contiguity of leads
4 Electrophysiology
5 Interpretation
5.1 Theory
5.2 Electrocardiogram grid
5.3 Rate and rhythm
5.4 Axis
5.5 Amplitudes and intervals
5.6 Ischemia and infarction
5.7 Artifacts
6 Diagnosis
7 History
8 See also
9 Notes
10 References
11 External links
Medical uses[edit]
A 12-lead ECG of a 26-year-old male with an incomplete RBBB
The overall goal of performing an ECG is to obtain information about the structure and function of the heart. Medical uses for this information are varied and generally need knowledge of the structure and/or function of the heart to be interpreted. Some indications for performing an ECG include:
Suspected myocardial infarction (heart attack) or chest pain
ST elevated myocardial infarction (STEMI)[9]
non-ST elevated myocardial infarction (NSTEMI)[10]
Suspected pulmonary embolism or shortness of breath
A third heart sound, fourth heart sound, a cardiac murmur[11] or other findings suggestive of a structural heart disease
Perceived arrhythmia either by pulse or palpitations
Monitoring of known cardiac arrhythmias
Fainting or collapse[11]
Seizures[11]
Monitoring the effects of a medication on the heart (e.g. drug-induced QT prolongation)
Assessing severity of electrolyte abnormalities, such as hyperkalemia
Hypertrophic cardiomyopathy screening in adolescents as part of a sports physical out of concern for sudden cardiac death (varies by country)
Perioperative monitoring in which any form of anesthesia is involved (e.g. monitored anesthesia care, general anesthesia); typically both intraoperative and postoperative
As a part of a preoperative assessment some time before a surgical procedure (especially for those with known cardiovascular disease or who are undergoing invasive, cardiac, vascular or pulmonary procedures, or who will receive general anesthesia)
Cardiac stress testing
Computed tomography angiography (CTA) and magnetic resonance angiography (MRA) of the heart (ECG is used to "gate" the scanning so that the anatomical position of the heart is steady)
Biotelemetry of patients for any of the above reasons and such monitoring can include internal and external defibrillators and pacemakers
The United States Preventive Services Task Force does not recommend an ECG for routine screening in patients without symptoms and those at low risk for coronary artery disease.[12][13] This is because an ECG may falsely indicate the existence of a problem, leading to misdiagnosis, the recommendation of invasive procedures, or overtreatment. However, persons employed in certain critical occupations, such as aircraft pilots,[14] may be required to have an ECG as part of their routine health evaluations.
Continuous ECG monitoring is used to monitor critically ill patients, patients undergoing general anesthesia,[11] and patients who have an infrequently occurring cardiac arrhythmia that would unlikely be seen on a conventional ten-second ECG.
In the United States, a 12-lead ECG is commonly performed by specialized technicians that may be certified as electrocardiogram technicians. ECG interpretation is a component of many healthcare fields (nurses and physicians and cardiac surgeons being the most obvious), but anyone trained to interpret an ECG is free to do so. However, "official" interpretation is performed by a cardiologist. Certain fields such as anesthesia utilize continuous ECG monitoring, and knowledge of interpreting ECGs is crucial to their jobs.
One additional form of ECG is used in clinical cardiac electrophysiology in which a catheter is used to measure the electrical activity. The catheter is inserted through the femoral vein and can have several electrodes along its length to record the direction of electrical activity from within the heart.
Evidence does not support the use of ECGs among those without symptoms or at low risk of cardiovascular disease as an effort for prevention.[15]
Electrocardiographs[edit]
An electrocardiograph with integrated display and keyboard on a wheeled cart
An electrocardiograph is a machine that is used to perform electrocardiography, and produces the electrocardiogram. The first electrocardiographs are discussed later and are electrically primitive compared to today's machines.
The fundamental component to an ECG is the instrumentation amplifier, which is responsible for taking the voltage difference between leads (see below) and amplifying the signal. ECG voltages measured across the body are on the order of hundreds of microvolts up to 1 millivolt (the small square on a standard ECG is 100 microvolts). This low voltage necessitates a low noise circuit and instrumentation amplifiers.
Early ECGs were constructed with analog electronics and the signal could drive a motor to print the signal on paper. Today, electrocardiographs use analog-to-digital converters to convert to a digital signal that can then be manipulated with digital electronics. This permits digital recording of ECGs and use on computers.
There are other components to the ECG:[16]
Safety features that include voltage protection for the patient and operator. Since the machines are powered by mains power, it is conceivable that either person could be subjected to voltage capable of causing death. Additionally, the heart is sensitive to the AC frequencies typically used for mains power (50 or 60 Hz).
Defibrillation protection: any ECG used in healthcare may be attached to a person who requires defibrillation and the ECG needs to protect itself from this source of energy.
Electrostatic discharge is similar to defibrillation discharge and requires voltage protection up to 18,000 volts.
Additionally circuitry called the right leg driver can be used to reduce common-mode interference (typically the 50 or 60 Hz mains power).
The typical design for a portable ECG is a combined unit that includes a screen, keyboard, and printer on a small wheeled cart. The unit connects to a long cable that branches to each lead and attaches to a conductive pad on the patient.
The ECG may include a rhythm analysis algorithm that produces a computerized interpretation of the ECG. The results from these algorithms are considered "preliminary" until verified and/or modified by someone trained in interpreting ECGs. Included in this analysis is computation of common parameters that include PR interval, QT interval, corrected QT (QTc) interval, PR axis, QRS axis, and more. Earlier designs recorded each lead sequentially but current designs employ circuits that can record all leads simultaneously. The former introduces problems in interpretation since there may be beat-to-beat changes in the rhythm, which makes it unwise to compare across beats.
More recent advancements in electrocardiography include work in diminishing the size of the unit to make it more portable and therefore more accessible to larger groups of patients. To achieve this, these smaller devices rely on only two electrodes which together deliver "lead I" of the standard ECG.[17]
Electrodes and leads[edit]
Proper placement of the limb electrodes. The limb electrodes can be far down on the limbs or close to the hips/shoulders as long as they are placed symmetrically.[18]
Placement of the precordial electrodes
Electrodes are the actual conductive pads attached to the body surface. Any pair of electrodes can measure the electrical potential difference between the two corresponding locations of attachment. Such a pair forms a lead. However, "leads" can also be formed between a physical electrode and a virtual electrode, known as the Wilson's central terminal, whose potential is defined as the average potential measured by three limb electrodes that are attached to the right arm, the left arm, and the left foot, respectively.
Commonly, 10 electrodes attached to the body are used to form 12 ECG leads, with each lead measuring a specific electrical potential difference (as listed in the table below).[citation needed]
Leads are broken down into three types: limb; augmented limb; and precordial or chest. The 12-lead ECG has a total of three limb leads and three augmented limb leads arranged like spokes of a wheel in the coronal plane (vertical), and six precordial leads or chest leads that lie on the perpendicular transverse plane (horizontal).[citation needed]
In medical settings, the term leads is also sometimes used to refer to the electrodes themselves, although this is technically incorrect. This misuse of terminology can be the source of confusion.[citation needed]
The 10 electrodes in a 12-lead ECG are listed below.[19]
Electrode name
Electrode placement
RA On the right arm, avoiding thick muscle.
LA In the same location where RA was placed, but on the left arm.
RL On the right leg, lower end of inner aspect of calf muscle. (Avoid bony prominences)
LL In the same location where RL was placed, but on the left leg.
V1 In the fourth intercostal space (between ribs 4 and 5) just to the right of the sternum (breastbone).
V2 In the fourth intercostal space (between ribs 4 and 5) just to the left of the sternum.
V3 Between leads V2 and V4.
V4 In the fifth intercostal space (between ribs 5 and 6) in the mid-clavicular line.
V5 Horizontally even with V4, in the left anterior axillary line.
V6 Horizontally even with V4 and V5 in the mid-axillary line.
Two types of electrodes in common use are a flat paper-thin sticker and a self-adhesive circular pad. The former are typically used in a single ECG recording while the latter are for continuous recordings as they stick longer. Each electrode consists of an electrically conductive electrolyte gel and a silver/silver chloride conductor.[20] The gel typically contains potassium chloride – sometimes silver chloride as well – to permit electron conduction from the skin to the wire and to the electrocardiogram.
The common virtual electrode, known as the Wilson's central terminal (VW), is produced by averaging the measurements from the electrodes RA, LA, and LL to give an average potential of the body:
V W = 1 3 ( R A + L A + L L ) {\displaystyle V_{W}={\frac {1}{3}}(RA+LA+LL)}
In a 12-lead ECG, all leads except the limb leads are unipolar (aVR, aVL, aVF, V1, V2, V3, V4, V5, and V6). The measurement of a voltage requires two contacts and so, electrically, the unipolar leads are measured from the common lead (negative) and the unipolar lead (positive). This averaging for the common lead and the abstract unipolar lead concept makes for a more challenging understanding and is complicated by sloppy usage of "lead" and "electrode."
Limb leads[edit]
The limb leads and augmented limb leads (Wilson's central terminal is used as the negative pole for the latter in this representation)
Leads I, II and III are called the limb leads. The electrodes that form these signals are located on the limbs – one on each arm and one on the left leg.[21][22][23] The limb leads form the points of what is known as Einthoven's triangle.[24]
Lead I is the voltage between the (positive) left arm (LA) electrode and right arm (RA) electrode:
I = L A − R A {\displaystyle I=LA-RA}
Lead II is the voltage between the (positive) left leg (LL) electrode and the right arm (RA) electrode:
I I = L L − R A {\displaystyle II=LL-RA}
Lead III is the voltage between the (positive) left leg (LL) electrode and the left arm (LA) electrode:
I I I = L L − L A {\displaystyle III=LL-LA}
Augmented limb leads[edit]
Leads aVR, aVL, and aVF are the augmented limb leads. They are derived from the same three electrodes as leads I, II, and III, but they use Goldberger's central terminal as their negative pole. Goldberger's central terminal is a combination of inputs from two limb electrodes, with a different combination for each augmented lead. It is referred to immediately below as "the negative pole."
Lead augmented vector right (aVR) has the positive electrode on the right arm. The negative pole is a combination of the left arm electrode and the left leg electrode:
a V R = R A − 1 2 ( L A + L L ) = 3 2 ( R A − V W ) {\displaystyle aVR=RA-{\frac {1}{2}}(LA+LL)={\frac {3}{2}}(RA-V_{W})}
Lead augmented vector left (aVL) has the positive electrode on the left arm. The negative pole is a combination of the right arm electrode and the left leg electrode:
a V L = L A − 1 2 ( R A + L L ) = 3 2 ( L A − V W ) {\displaystyle aVL=LA-{\frac {1}{2}}(RA+LL)={\frac {3}{2}}(LA-V_{W})}
Lead augmented vector foot (aVF) has the positive electrode on the left leg. The negative pole is a combination of the right arm electrode and the left arm electrode:
a V F = L L − 1 2 ( R A + L A ) = 3 2 ( L L − V W ) {\displaystyle aVF=LL-{\frac {1}{2}}(RA+LA)={\frac {3}{2}}(LL-V_{W})}
Together with leads I, II, and III, augmented limb leads aVR, aVL, and aVF form the basis of the hexaxial reference system, which is used to calculate the heart's electrical axis in the frontal plane.
Precordial leads[edit]
The precordial leads lie in the transverse (horizontal) plane, perpendicular to the other six leads. The six precordial electrodes act as the positive poles for the six corresponding precordial leads: (V1, V2, V3, V4, V5 and V6). Wilson's central terminal is used as the negative pole.
Specialized leads[edit]
Additional electrodes may rarely be placed to generate other leads for specific diagnostic purposes. Right-sided precordial leads may be used to better study pathology of the right ventricle or for dextrocardia (and are denoted with an R (e.g., V5R). Posterior leads (V7 to V9) may be used to demonstrate the presence of a posterior myocardial infarction. A Lewis lead (requiring an electrode at the right sternal border in the second intercostal space) can be used to study pathological rhythms arising in the right atrium.
An esophogeal lead can be inserted to a part of the esophagus where the distance to the posterior wall of the left atrium is only approximately 5–6 mm (remaining constant in people of different age and weight).[25] An esophageal lead avails for a more accurate differentiation between certain cardiac arrhythmias, particularly atrial flutter, AV nodal reentrant tachycardia and orthodromic atrioventricular reentrant tachycardia.[26] It can also evaluate the risk in people with Wolff-Parkinson-White syndrome, as well as terminate supraventricular tachycardia caused by re-entry.[26]
An intracardiac electrogram (ICEG) is essentially an ECG with some added intracardiac leads (that is, inside the heart). The standard ECG leads (external leads) are I, II, III, aVL, V1, and V6. Two to four intracardiac leads are added via cardiac catheterization. The word "electrogram" (EGM) without further specification usually means an intracardiac electrogram.
Lead locations on an ECG report[edit]
A standard 12-lead ECG report (an electrocardiograph) shows a 2.5 second tracing of each of the twelve leads. The tracings are most commonly arranged in a grid of four columns and three rows. the first column is the limb leads (I, II, and III), the second column is the augmented limb leads (aVR, aVL, and aVF), and the last two columns are the precordial leads (V1 to V6). Additionally, a rhythm strip may be included as a fourth or fifth row.
The timing across the page is continuous and not tracings of the 12 leads for the same time period. In other words, if the output were traced by needles on paper, each row would switch which leads as the paper is pulled under the needle. For example, the top row would first trace lead I, then switch to lead aVR, then switch to V1, and then switch to V4 and so none of these four tracings of the leads are from the same time period as they are traced in sequence through time.
Contiguity of leads[edit]
Diagram showing the contiguous leads in the same color in the standard 12-lead layout
Each of the 12 ECG leads records the electrical activity of the heart from a different angle, and therefore align with different anatomical areas of the heart. Two leads that look at neighboring anatomical areas are said to be contiguous.
Category
Leads
Activity
Inferior leads Leads II, III and aVF Look at electrical activity from the vantage point of the inferior surface (diaphragmatic surface of heart)
Lateral leads I, aVL, V5 and V6 Look at the electrical activity from the vantage point of the lateral wall of left ventricle
Septal leads V1 and V2 Look at electrical activity from the vantage point of the septal surface of the heart (interventricular septum)
Anterior leads V3 and V4 Look at electrical activity from the vantage point of the anterior wall of the right and left ventricles (Sternocostal surface of heart)
In addition, any two precordial leads next to one another are considered to be contiguous. For example, though V4 is an anterior lead and V5 is a lateral lead, they are contiguous because they are next to one another.
Electrophysiology[edit]
Main article: Cardiac electrophysiology
The formal study of the electrical conduction system of the heart is called cardiac electrophysiology (EP). An electrophysiology study involves a formal study of the conduction system and can be done for various reasons. During such a study, catheters are used to access the heart and some of these catheters include electrodes that can be placed anywhere in the heart to record the electrical activity from within the heart. Some catheters contain several electrodes and can record the propagation of electrical activity.
Interpretation[edit]
Interpretation of the ECG is fundamentally about understanding the electrical conduction system of the heart. Normal conduction starts and propagates in a predictable pattern, and deviation from this pattern can be a normal variation or be pathological. An ECG does not equate with mechanical pumping activity of the heart, for example, pulseless electrical activity produces an ECG that should pump blood but no pulses are felt (and constitutes a medical emergency and CPR should be performed). Ventricular fibrillation produces an ECG but is too dysfunctional to produce a life-sustaining cardiac output. Certain rhythms are known to have good cardiac output and some are known to have bad cardiac output. Ultimately, an echocardiogram or other anatomical imaging modality is useful in assessing the mechanical function of the heart.
Like all medical tests, what constitutes "normal" is based on population studies. The heartrate range of between 60 and 100 beats per minute (bpm) is considered normal since data shows this to be the usual resting heart rate.
Theory[edit]
QRS is upright in a lead when its axis is aligned with that lead's vector
Schematic representation of a normal ECG
Interpretation of the ECG is ultimately that of pattern recognition. In order to understand the patterns found, it is helpful to understand the theory of what ECGs represent. The theory is rooted in electromagnetics and boils down to the four following points:
depolarization of the heart toward the positive electrode produces a positive deflection
depolarization of the heart away from the positive electrode produces a negative deflection
repolarization of the heart toward the positive electrode produces a negative deflection
repolarization of the heart away from the positive electrode produces a positive deflection
Thus, the overall direction of depolarization and repolarization produces a vector that produces positive or negative deflection on the ECG depending on which lead it points to. For example, depolarizing from right to left would produce a positive deflection in lead I because the two vectors point in the same direction. In contrast, that same depolarization would produce minimal deflection in V1 and V2 because the vectors are perpendicular and this phenomenon is called isoelectric.
Normal rhythm produces four entities – a P wave, a QRS complex, a T wave, and a U wave – that each have a fairly unique pattern.
The P wave represents atrial depolarization.
The QRS complex represents ventricular depolarization.
The T wave represents ventricular repolarization.
The U wave represents papillary muscle repolarization.
However, the U wave is not typically seen and its absence is generally ignored. Changes in the structure of the heart and its surroundings (including blood composition) change the patterns of these four entities.
Electrocardiogram grid[edit]
ECGs are normally printed on a grid. The horizontal axis represents time and the vertical axis represents voltage. The standard values on this grid are shown in the adjacent image:
A small box is 1 mm × 1 mm and represents 0.1 mV × 0.04 seconds.
A large box is 5 mm × 5 mm and represents 0.5 mV × 0.20 seconds.
The "large" box is represented by a heavier line weight than the small boxes.
Not all aspects of an ECG rely on precise recordings or having a known scaling of amplitude or time. For example, determining if the tracing is a sinus rhythm only requires feature recognition and matching, and not measurement of amplitudes or times (i.e., the scale of the grids are irrelevant). An example to the contrary, the voltage requirements of left ventricular hypertrophy require knowing the grid scale.
Rate and rhythm[edit]
In a normal heart, the heart rate is the rate in which the sinoatrial node depolarizes as it is the source of depolarization of the heart. Heart rate, like other vital signs like blood pressure and respiratory rate, change with age. In adults, a normal heart rate is between 60 and 100 bpm (normocardic) where in children it is higher. A heart rate less than normal is called bradycardia (<60 in adults) and higher than normal is tachycardia (>100 in adults). A complication of this is when the atria and ventricles are not in synchrony and the "heart rate" must be specified as atrial or ventricular (e.g., the ventricular rate in ventricular fibrillation is 300–600 bpm, whereas the atrial rate can be normal [60–100] or faster [100–150]).
In normal resting hearts, the physiologic rhythm of the heart is normal sinus rhythm (NSR). Normal sinus rhythm produces the prototypical pattern of P wave, QRS complex, and T wave. Generally, deviation from normal sinus rhythm is considered a cardiac arrhythmia. Thus, the first question in interpreting an ECG is whether or not there is a sinus rhythm. A criterion for sinus rhythm is that P waves and QRS complexes appear 1-to-1, thus implying that the P wave causes the QRS complex.
Once sinus rhythm is established, or not, the second question is the rate. For a sinus rhythm this is either the rate of P waves or QRS complexes since they are 1-to-1. If the rate is too fast then it is sinus tachycardia and if it is too slow then it is sinus bradycardia.
If it is not a sinus rhythm, then determining the rhythm is necessary before proceeding with further interpretation. Some arrhythmias with characteristic findings:
Absent P waves with "irregularly irregular" QRS complexes is the hallmark of atrial fibrillation
A "saw tooth" pattern with QRS complexes is the hallmark of atrial flutter
Sine wave pattern is the hallmark of ventricular flutter
Absent P waves with wide QRS complexes and a fast heart rate is ventricular tachycardia
Determination of rate and rhythm is necessary in order to make sense of further interpretation.
Axis[edit]
The heart has several axes, but the most common by far is the axis of the QRS complex (references to "the axis" imply the QRS axis). Each axis can be computationally determined to result in a number representing degrees of deviation from zero, or it can be categorized into a few types.
The QRS axis is the general direction of the ventricular depolarization wavefront (or mean electrical vector) in the frontal plane. It is often sufficient to classify the axis as one of three types: normal, left deviated, or right deviated. Population data shows that a normal QRS axis is from −30° to 105°, with 0° being along lead I and positive being inferior and negative being superior (best understood graphically as the hexaxial reference system).[27] Beyond +105° is right axis deviation and beyond −30° is left axis deviation (the third quadrant of −90° to −180° is very rare and is an indeterminate axis). A shortcut for determining if the QRS axis is normal is if the QRS complex is mostly positive in lead I and lead II (or lead I and aVF if +90° is the upper limit of normal).
The normal QRS axis is generally down and to the left, following the anatomical orientation of the heart within the chest. An abnormal axis suggests a change in the physical shape and orientation of the heart or a defect in its conduction system that causes the ventricles to depolarize in an abnormal way.
Classification
Angle
Notes
Normal −30° to 105° Normal
Left axis deviation −30° to −90° May indicate left ventricular hypertrophy, left anterior fascicular block, or an old inferior STEMI
Right axis deviation +105° to +180° May indicate right ventricular hypertrophy, left posterior fascicular block, or an old lateral STEMI
Indeterminate axis +180° to −90° Rarely seen; considered an 'electrical no-man's land'
The extent of a normal axis can be +90° or 105° depending on the source.
Amplitudes and intervals[edit]
Animation of a normal ECG wave
All of the waves on an ECG tracing and the intervals between them have a predictable time duration, a range of acceptable amplitudes (voltages), and a typical morphology. Any deviation from the normal tracing is potentially pathological and therefore of clinical significance.
For ease of measuring the amplitudes and intervals, an ECG is printed on graph paper at a standard scale: each 1 mm (one small box on the standard ECG paper) represents 40 milliseconds of time on the x-axis, and 0.1 millivolts on the y-axis.
Feature
Description
Pathology
Duration
P wave The P wave represents depolarization of the atria. Atrial depolarization spreads from the SA node towards the AV node, and from the right atrium to the left atrium. The P wave is typically upright in most leads except for aVR; an unusual P wave axis (inverted in other leads) can indicate an ectopic atrial pacemaker. If the P wave is of unusually long duration, it may represent atrial enlargement. Typically a large right atrium gives a tall, peaked P wave while a large left atrium gives a two-humped bifid P wave. <80 ms
PR interval The PR interval is measured from the beginning of the P wave to the beginning of the QRS complex. This interval reflects the time the electrical impulse takes to travel from the sinus node through the AV node. A PR interval shorter than 120 ms suggests that the electrical impulse is bypassing the AV node, as in Wolf-Parkinson-White syndrome. A PR interval consistently longer than 200 ms diagnoses first degree atrioventricular block. The PR segment (the portion of the tracing after the P wave and before the QRS complex) is typically completely flat, but may be depressed in pericarditis. 120 to 200 ms
QRS complex The QRS complex represents the rapid depolarization of the right and left ventricles. The ventricles have a large muscle mass compared to the atria, so the QRS complex usually has a much larger amplitude than the P wave. If the QRS complex is wide (longer than 120 ms) it suggests disruption of the heart's conduction system, such as in LBBB, RBBB, or ventricular rhythms such as ventricular tachycardia. Metabolic issues such as severe hyperkalemia, or tricyclic antidepressant overdose can also widen the QRS complex. An unusually tall QRS complex may represent left ventricular hypertrophy while a very low-amplitude QRS complex may represent a pericardial effusion or infiltrative myocardial disease. 80 to 100 ms
J-point The J-point is the point at which the QRS complex finishes and the ST segment begins. The J-point may be elevated as a normal variant. The appearance of a separate J wave or Osborn wave at the J-point is pathognomonic of hypothermia or hypercalcemia.[28]
ST segment The ST segment connects the QRS complex and the T wave; it represents the period when the ventricles are depolarized. It is usually isoelectric, but may be depressed or elevated with myocardial infarction or ischemia. ST depression can also be caused by LVH or digoxin. ST elevation can also be caused by pericarditis, Brugada syndrome, or can be a normal variant (J-point elevation).
T wave The T wave represents the repolarization of the ventricles. It is generally upright in all leads except aVR and lead V1. Inverted T waves can be a sign of myocardial ischemia, left ventricular hypertrophy, high intracranial pressure, or metabolic abnormalities. Peaked T waves can be a sign of hyperkalemia or very early myocardial infarction. 160 ms
Corrected QT interval (QTc) The QT interval is measured from the beginning of the QRS complex to the end of the T wave. Acceptable ranges vary with heart rate, so it must be corrected to the QTc by dividing by the square root of the RR interval. A prolonged QTc interval is a risk factor for ventricular tachyarrhythmias and sudden death. Long QT can arise as a genetic syndrome, or as a side effect of certain medications. An unusually short QTc can be seen in severe hypercalcemia. <440 ms
U wave The U wave is hypothesized to be caused by the repolarization of the interventricular septum. It normally has a low amplitude, and even more often is completely absent. If the U wave is very prominent, suspect hypokalemia, hypercalcemia or hyperthyroidism.[29]
Ischemia and infarction[edit]
Main article: Electrocardiography in myocardial infarction
Ischemia or non-ST elevation myocardial infarctions (non-STEMIs) may manifest as ST depression or inversion of T waves. It may also affect the high frequency band of the QRS.
ST elevation myocardial infarctions (STEMIs) have different characteristic ECG findings based on the amount of time elapsed since the MI first occurred. The earliest sign is hyperacute T waves, peaked T waves due to local hyperkalemia in ischemic myocardium. This then progresses over a period of minutes to elevations of the ST segment by at least 1 mm. Over a period of hours, a pathologic Q wave may appear and the T wave will invert. Over a period of days the ST elevation will resolve. Pathologic Q waves generally will remain permanently.[30]
The coronary artery that has been occluded can be identified in an STEMI based on the location of ST elevation. The left anterior descending (LAD) artery supplies the anterior wall of the heart, and therefore causes ST elevations in anterior leads (V1 and V2). The LCx supplies the lateral aspect of the heart and therefore causes ST elevations in lateral leads (I, aVL and V6). The right coronary artery (RCA) usually supplies the inferior aspect of the heart, and therefore causes ST elevations in inferior leads (II, III and aVF).
Artifacts[edit]
An ECG tracing is affected by patient motion. Some rhythmic motions (such as shivering or tremors) can create the illusion of cardiac arrhythmia.[31] Artifacts are distorted signals caused by a secondary internal or external sources, such as muscle movement or interference from an electrical device.[32][33]
Distortion poses significant challenges to healthcare providers,[32] who employ various techniques[34] and strategies to safely recognize[35] these false signals.[medical citation needed] Accurately separating the ECG artifact from the true ECG signal can have a significant impact on patient outcomes and legal liabilities.[36][unreliable medical source?]
Improper lead placement (for example, reversing two of the limb leads) has been estimated to occur in 0.4% to 4% of all ECG recordings,[37] and has resulted in improper diagnosis and treatment including unnecessary use of thrombolytic therapy.[38][39]
Diagnosis[edit]
Numerous diagnoses and findings can be made based upon electrocardiography, and many are discussed above. Overall, the diagnoses are made based on the patterns. For example, an "irregularly irregular" QRS complex without P waves is the hallmark of atrial fibrillation; however, other findings can be present as well, such as a bundle branch block that alters the shape of the QRS complexes. ECGs can be interpreted in isolation but should be applied – like all diagnostic tests – in the context of the patient. For example, an observation of peaked T waves is not sufficient to diagnose hyperkalemia; such a diagnosis should be verified by measuring the blood potassium level. Conversely, a discovery of hyperkalemia should be followed by an ECG for manifestations such as peaked T waves, widened QRS complexes, and loss of P waves. The following is an organized list of possible ECG-based diagnoses.
Rhythm disturbances or arrhythmias:
Atrial fibrillation and atrial flutter without rapid ventricular response
Premature atrial contraction (PACs) and premature ventricular contraction (PVCs)
Sinus arrhythmia
Sinus bradycardia and sinus tachycardia
Sinus pause and sinoatrial arrest
Sick sinus syndrome: bradycardia-tachycardia syndrome
Supraventricular tachycardia
Atrial fibrillation with rapid ventricular response
Atrial flutter with rapid ventricular response
AV nodal reentrant tachycardia
Atrioventricular reentrant tachycardia
Junctional ectopic tachycardia
Atrial tachycardia
Ectopic atrial tachycardia (unicentric)
Multifocal atrial tachycardia
Paroxysmal atrial tachycardia
Sinoatrial nodal reentrant tachycardia
Torsades de pointes (polymorphic ventricular tachycardia)
Wide complex tachycardia
Ventricular flutter
Ventricular fibrillation
Ventricular tachycardia (monomorphic ventricular tachycardia)
Pre-excitation syndrome
Lown–Ganong–Levine syndrome
Wolff–Parkinson–White syndrome
J wave (Osborn wave)
Heart block and conduction problems:
Aberration
Sinoatrial block: first, second, and third-degree
AV node
First-degree AV block
Second-degree AV block (Mobitz [Wenckebach] I and II)
Third-degree AV block or complete AV block
Right bundle
Incomplete right bundle branch block
Complete right bundle branch block (RBBB)
Left bundle
Complete left bundle branch block (LBBB)
Incomplete left bundle branch block
Left anterior fascicular block (LAFB)
Left posterior fascicular block (LPFB)
Bifascicular block (LAFB plus LPFB)
Trifascicular block (LAFP plus FPFB plus RBBB)
QT syndromes
Brugada syndrome
Short QT syndrome
Long QT syndromes, genetic and drug-induced
Right and left atrial abnormality
Electrolytes disturbances and intoxication:
Digitalis intoxication
Calcium: hypocalcemia and hypercalcemia
Potassium: hypokalemia and hyperkalemia
Ischemia and infarction:
Wellens' syndrome (LAD occlusion)
de Winter T waves (LAD occlusion) [40]
ST elevation and ST depression
High Frequency QRS changes
Myocardial infarction (heart attack)
Non-Q wave myocardial infarction
NSTEMI
STEMI
Sgarbossa's criteria for ischemia with a LBBB
Structural:
Acute pericarditis
Right and left ventricular hypertrophy
Right ventricular strain or S1Q3T3 (can be seen in pulmonary embolism)
History[edit]
An early commercial ECG device (1911)
ECG from 1957
The etymology of the word is derived from the Greek electro, because it is related to electrical activity, kardia, Greek for heart, and graph, a Greek root meaning "to write".
Alexander Muirhead is reported to have attached wires to a feverish patient's wrist to obtain a record of the patient's heartbeat in 1872 at St Bartholomew's Hospital.[41] Another early pioneer was Augustus Waller, of St Mary's Hospital in London.[42] His electrocardiograph machine consisted of a Lippmann capillary electrometer fixed to a projector. The trace from the heartbeat was projected onto a photographic plate that was itself fixed to a toy train. This allowed a heartbeat to be recorded in real time.
An initial breakthrough came when Willem Einthoven, working in Leiden, the Netherlands, used the string galvanometer (the first practical electrocardiograph) he invented in 1901.[43] This device was much more sensitive than both the capillary electrometer Waller used and the string galvanometer that had been invented separately in 1897 by the French engineer Clément Ader.[44] Einthoven had previously, in 1895, assigned the letters P, Q, R, S, and T to the deflections in the theoretical waveform he created using equations which corrected the actual waveform obtained by the capillary electrometer to compensate for the imprecision of that instrument. Using letters different from A, B, C, and D (the letters used for the capillary electrometer's waveform) facilitated comparison when the uncorrected and corrected lines were drawn on the same graph.[45] Einthoven probably chose the initial letter P to follow the example set by Descartes in geometry.[45] When a more precise waveform was obtained using the string galvanometer, which matched the corrected capillary electrometer waveform, he continued to use the letters P, Q, R, S, and T,[45] and these letters are still in use today. Einthoven also described the electrocardiographic features of a number of cardiovascular disorders. In 1924, he was awarded the Nobel Prize in Medicine for his discovery.[46]
By 1927, General Electric had developed a portable apparatus that could produce electrocardiograms without the use of the string galvanometer. This device instead combined amplifier tubes similar to those used in a radio with an internal lamp and a moving mirror that directed the tracing of the electric pulses onto film.[47]
In 1937, Taro Takemi invented a new portable electrocardiograph machine.[48]
Though the basic principles of that era are still in use today, many advances in electrocardiography have been made over the years. Instrumentation has evolved from a cumbersome laboratory apparatus to compact electronic systems that often include computerized interpretation of the electrocardiogram.[49]
In September 2018, Apple Inc., introduced the Apple Watch Series 4, with a built-in titanium electrode in the digital crown and the sapphire crystal electronic heart sensor, which allows the watch to give a single lead electrocardiogram using only the watch interface.[50]
See also[edit]
Electrical conduction system of the heart
Electrogastrogram
Electropalatography
Electroretinography
Heart rate monitor
Emergency medicine
Notes[edit]
Jump up ^ The version with '-K-', more commonly used in American English than in British English, is an early-20th-century loanword from the German acronym EKG for Elektrokardiogramm (electrocardiogram),[1] which reflects that German physicians were pioneers in the field at the time. Today AMA style and – under its stylistic influence – most American medical publications use ECG instead of EKG.[2] The German term Elektrokardiogramm as well as the English equivalent, electrocardiogram, consist of the New Latin/international scientific vocabulary elements elektro- (cognate electro-) and kardi- (cognate 'cardi-'), the latter from Greek kardia (heart).[3] The '-K-' version is more often retained under circumstances where there may be verbal confusion between ECG and EEG (electroencephalography) due to similar pronunciation.
References[edit]
Jump up ^ EKG. Oxford Online Dictionaries
Jump up ^ "15.3.1 Electrocardiographic Terms", AMA Manual of Style, American Medical Association
Jump up ^ "Merriam-Webster's Collegiate Dictionary". Merriam-Webster.
Jump up ^ Lilly, Leonard S, ed. (2016). Pathophysiology of Heart Disease: A Collaborative Project of Medical Students and Faculty (sixth ed.). Lippincott Williams & Wilkins. p. 74. ISBN 978-1451192759.
Jump up ^ Aswini Kumar MD. "ECG- simplified". LifeHugger. Retrieved 11 February 2010.
^ Jump up to: a b Lilly 2016, pp. 80.
Jump up ^ Walraven, G. (2011). Basic arrhythmias (7th ed.), pp. 1–11
Jump up ^ Braunwald E. (ed) (1997), Heart Disease: A Textbook of Cardiovascular Medicine, Fifth Edition, p. 108, Philadelphia, W.B. Saunders Co.. ISBN 0-7216-5666-8.
Jump up ^ "What is a STEMI? - ECG Medical Training". ECG Medical Training. 2015-06-24. Retrieved 2018-06-24.
Jump up ^ "What is NSTEMI? What You NEED to Know • MyHeart". MyHeart. 2015-04-30. Retrieved 2018-06-24.
^ Jump up to: a b c d Masters, Jo; Bowden, Carole; Martin, Carole (2003). Textbook of veterinary medical nursing. Oxford: Butterworth-Heinemann. p. 244. ISBN 978-0-7506-5171-4.
Jump up ^ Moyer VA (2 October 2012). "Screening for coronary heart disease with electrocardiography: U.S. Preventive Services Task Force recommendation statement". Annals of Internal Medicine. 157 (7): 512–18. doi:10.7326/0003-4819-157-7-201210020-00514. PMID 22847227.
Jump up ^ Consumer Reports; American Academy of Family Physicians; ABIM Foundation (April 2012), "EKGs and exercise stress tests: When you need them for heart disease — and when you don't" (PDF), Choosing Wisely, Consumer Reports, retrieved 14 August 2012
Jump up ^ "Summary of Medical Standards" (PDF). U.S. Federal Aviation Administration. 2006. Retrieved 27 December 2013.
Jump up ^ US Preventive Services Task, Force.; Curry, SJ; Krist, AH; Owens, DK; Barry, MJ; Caughey, AB; Davidson, KW; Doubeni, CA; Epling JW, Jr; Kemper, AR; Kubik, M; Landefeld, CS; Mangione, CM; Silverstein, M; Simon, MA; Tseng, CW; Wong, JB (12 June 2018). "Screening for Cardiovascular Disease Risk With Electrocardiography: US Preventive Services Task Force Recommendation Statement". JAMA. 319 (22): 2308–2314. doi:10.1001/jama.2018.6848. PMID 29896632.
Jump up ^ "Mitigation Strategies for ECG Design Challenges" (PDF). Analog Devices. Retrieved 24 April 2016.
Jump up ^ "FDA approves AliveCor heart monitor". Techcrunch. Retrieved 2018-08-25.
Jump up ^ Macfarlane, P.W.; Coleman (1995). "Resting 12-Lead Electrode" (PDF). Society for Cardiological Science and Technology. Retrieved 21 October 2017.
Jump up ^ "12-Lead ECG Placement Guide with Illustrations". Cables and Sensors. Retrieved 11 July 2017.
Jump up ^ Kavuru, Madhav S.; Vesselle, Hubert; Thomas, Cecil W. (1987). Advances in Body Surface Potential Mapping (BSPM) Instrumentation. Pediatric and Fundamental Electrocardiography. Developments in Cardiovascular Medicine. 56. pp. 315–327. doi:10.1007/978-1-4613-2323-5_15. ISBN 978-1-4612-9428-3. ISSN 0166-9842.
Jump up ^ Sensors, Cables and. "12-Lead ECG Placement Guide with Illustrations | Cables and Sensors". Cables and Sensors. Retrieved 2017-10-21.
Jump up ^ "Limb Leads – ECG Lead Placement – Normal Function of the Heart – Cardiology Teaching Package – Practice Learning – Division of Nursing – The University of Nottingham". Nottingham.ac.uk. Retrieved 15 August 2009.
Jump up ^ "Lesson 1: The Standard 12 Lead ECG". Library.med.utah.edu. Archived from the original on 22 March 2009. Retrieved 15 August 2009.
Jump up ^ Jin, Benjamin E.; Wulff, Heike; Widdicombe, Jonathan H.; Zheng, Jie; Bers, Donald M.; Puglisi, Jose L. (December 2012). "A simple device to illustrate the Einthoven triangle". Advances in Physiology Education. 36 (4): 319–24. doi:10.1152/advan.00029.2012. ISSN 1043-4046. PMC 3776430. PMID 23209014.
Jump up ^ Meigas, K; Kaik, J; Anier, A (2008). "Device and methods for performing transesophageal stimulation at reduced pacing current threshold". Estonian Journal of Engineering. 57 (2): 154. doi:10.3176/eng.2008.2.05.
^ Jump up to: a b Pehrson, Steen M.; Blomströ-Lundqvist, Carina; Ljungströ, Erik; Blomströ, Per (1994). "Clinical value of transesophageal atrial stimulation and recording in patients with arrhythmia-related symptoms or documented supraventricular tachycardia-correlation to clinical history and invasive studies". Clinical Cardiology. 17 (10): 528–34. doi:10.1002/clc.4960171004. PMID 8001299.
Jump up ^ Surawicz, Borys; Knillans, Timothy (2008). Chou's electrocardiography in clinical practice : adult and pediatric (6th ed.). Philadelphia, PA: Saunders/Elsevier. p. 12. ISBN 978-1416037743.
Jump up ^ Otero J, Lenihan DJ (2000). "The "normothermic" Osborn wave induced by severe hypercalcemia". Tex Heart Inst J. 27 (3): 316–17. PMC 101092. PMID 11093425.
Jump up ^ Houghton, Andrew R; Gray, David (2012). Making Sense of the ECG, Third Edition. Hodder Education. p. 214. ISBN 978-1-4441-6654-5.
Jump up ^ Alpert JS, Thygesen K, Antman E, Bassand JP (2000). "Myocardial infarction redefined – a consensus document of The Joint European Society of Cardiology/American College of Cardiology Committee for the redefinition of myocardial infarction". J Am Coll Cardiol. 36 (3): 959–69. doi:10.1016/S0735-1097(00)00804-4. PMID 10987628.
Jump up ^ Segura-Sampedro, Juan José; Parra-López, Loreto; Sampedro-Abascal, Consuelo; Muñoz-Rodríguez, Juan Carlos (2015). "Atrial flutter EKG can be useless without the proper electrophysiological basis". International Journal of Cardiology. 179: 68–69. doi:10.1016/j.ijcard.2014.10.076. PMID 25464416.
^ Jump up to: a b Takla, George; Petre, John H.; Doyle, D John; Horibe, Mayumi; Gopakumaran, Bala (2006). "The Problem of Artifacts in Patient Monitor Data During Surgery: A Clinical and Methodological Review". Anesthesia & Analgesia. 103 (5): 1196–204. doi:10.1213/01.ane.0000247964.47706.5d. PMID 17056954.
Jump up ^ Kligfield, Paul; Gettes, Leonard S.; Bailey, James J.; Childers, Rory; Deal, Barbara J.; Hancock, E. William; van Herpen, Gerard; Kors, Jan A.; Macfarlane, Peter (2007-03-13). "Recommendations for the standardization and interpretation of the electrocardiogram: part I: The electrocardiogram and its technology: a scientific statement from the American Heart Association Electrocardiography and Arrhythmias Committee, Council on Clinical Cardiology; the American College of Cardiology Foundation; and the Heart Rhythm Society: endorsed by the International Society for Computerized Electrocardiology". Circulation. 115 (10): 1306–24. doi:10.1161/CIRCULATIONAHA.106.180200. PMID 17322457.
Jump up ^ "Minimizing ECG Artifact" (PDF). Physio-Control. Physio-Control, Inc., Redmond WA. 2015. Retrieved 21 October 2017.
Jump up ^ Jafary, Fahim H (2007). "The "incidental" episode of ventricular fibrillation: A case report". Journal of Medical Case Reports. 1: 72. doi:10.1186/1752-1947-1-72. PMC 2000884. PMID 17760955.
Jump up ^ Mangalmurti, Sandeep; Seabury, Seth A.; Chandra, Amitabh; Lakdawalla, Darius; Oetgen, William J.; Jena, Anupam B. (2014). "Medical professional liability risk among US cardiologists". American Heart Journal. 167 (5): 690–96. doi:10.1016/j.ahj.2014.02.007. PMC 4153384. PMID 24766979.
Jump up ^ Incorrect electrode cable connection during electrocardiographic recording (2007) Velislav N. Batchvarov, Marek Malik, A. John Camm, Europace, Oct 2007
Jump up ^ Chanarin, N., Caplin, J., & Peacock, A. (1990). "Pseudo reinfarction": a consequence of electrocardiogram lead transposition following myocardial infarction. Clinical cardiology, 13(9), 668–69.
Jump up ^ Guijarro-Morales A., Gil-Extremera B., Maldonado-Martín A. (1991). "ECG diagnostic errors due to improper connection of the right arm and leg cables". International Journal of Cardiology. 30 (2): 233–35. doi:10.1016/0167-5273(91)90103-v. CS1 maint: Multiple names: authors list (link)
Jump up ^ de Winter, Robert (6 Nov 2008). "A New ECG Sign of Proximal LAD Occlusion". NEJM. 359 (19): 2071–3. doi:10.1056/NEJMc0804737. PMID 18987380.
Jump up ^ Ronald M. Birse,rev. Patricia E. Knowlden Oxford Dictionary of National Biography 2004 (Subscription required) – (original source is his biography written by his wife – Elizabeth Muirhead. Alexandernn Muirhead 1848–1920. Oxford, Blackwell: privately printed 1926.)
Jump up ^ Waller AD (1887). "A demonstration on man of electromotive changes accompanying the heart's beat". J Physiol. 8 (5): 229–34. doi:10.1113/jphysiol.1887.sp000257. PMC 1485094. PMID 16991463.
Jump up ^ Rivera-Ruiz M, Cajavilca C, Varon J (29 September 1927). "Einthoven's String Galvanometer: The First Electrocardiograph". Texas Heart Institute Journal / from the Texas Heart Institute of St. Luke's Episcopal Hospital, Texas Children's Hospital. 35 (2): 174–78. PMC 2435435. PMID 18612490.
Jump up ^ Interwoven W (1901). "Un nouveau galvanometre". Arch Neerl Sc Ex Nat. 6: 625.
^ Jump up to: a b c Hurst JW (3 November 1998). "Naming of the Waves in the ECG, With a Brief Account of Their Genesis". Circulation. 98 (18): 1937–42. doi:10.1161/01.CIR.98.18.1937. PMID 9799216.
Jump up ^ Cooper JK (1986). "Electrocardiography 100 years ago. Origins, pioneers, and contributors". N Engl J Med. 315 (7): 461–64. doi:10.1056/NEJM198608143150721. PMID 3526152.
Jump up ^ Blackford, John M., MD (1 May 1927). "Electrocardiography: A Short Talk Before the Staff of the Hospital". Clinics of the Virginia Mason Hospital. 6 (1): 28–34. CS1 maint: Multiple names: authors list (link)
Jump up ^ "Dr. Taro Takemi". Takemi Program in International Health. 2012-08-27. Retrieved 2017-10-21.
Jump up ^ Mark, Jonathan B. (1998). Atlas of cardiovascular monitoring. New York: Churchill Livingstone. ISBN 978-0-443-08891-9.
Jump up ^ "Apple Watch Series 4 - Health". Apple. Retrieved 2018-09-27.
External links[edit]
Wikimedia Commons has media related to ECG.
The whole ECG course on 1 A4 paper from ECGpedia, a wiki encyclopedia for a course on interpretation of ECG
Wave Maven – a large database of practice ECG questions provided by Beth Israel Deaconess Medical Center
PysioBank – a free scientific database with physiologic signals (here ecg)
EKG Academy – free EKG lectures, drills and quizzes
ECG Learning Center created by Eccles Health Sciences Library at University of Utah
show
v
t
e
Emergency medicine
Emergency medicine
Emergency department
Emergency medical services
Emergency nursing
Emergency psychiatry
Golden hour
Medical emergency
International emergency medicine
Pediatric emergency medicine
Pre-hospital emergency medicine
Major trauma
Trauma center
Triage
Equipment
Bag valve mask (BVM)
Chest tube
Defibrillation (AED
ICD)
Electrocardiogram (ECG/EKG)
Intraosseous infusion (IO)
Intravenous therapy (IV)
Tracheal intubation
Laryngeal tube
Combitube
Nasopharyngeal airway (NPA)
Oropharyngeal airway (OPA)
Pocket mask
Drugs
Adenosine
Amiodarone
Atropine
Dopamine
Epinephrine / Adrenaline
Naloxone
Magnesium sulfate
Sodium bicarbonate
Organisations
International Federation for Emergency Medicine (International Conference on Emergency Medicine)
American College of Emergency Physicians
Australasian College for Emergency Medicine
Canadian Association of Emergency Physicians
Royal College of Emergency Medicine
European Society of Emergency Medicine
Asian Society for Emergency Medicine
Courses / Life support
Acute Care of at-Risk Newborns (ACoRN)
Advanced cardiac life support (ACLS)
Advanced trauma life support (ATLS)
Basic life support (BLS)
Cardiopulmonary resuscitation (CPR)
Care of the Critically Ill Surgical Patient (CCrISP)
First aid
Neonatal Resuscitation Program (NRP)
Pediatric Advanced Life Support (PALS)
Scoring systems
NACA score
Injury Severity Score
Book:Emergency medicine
Category:Emergency medicine
show
v
t
e
Surgery and other procedures involving the heart (ICD-9-CM V3 35–37+89.4+99.6, ICD-10-PCS 02)
Surgery and IC
Heart valves
and septa
Valve repair
Valvulotomy
Mitral valve repair
Valvuloplasty
aortic
mitral
Valve replacement
Aortic valve repair
Aortic valve replacement
Ross procedure
Percutaneous aortic valve replacement
Mitral valve replacement
production of septal defect in heart
enlargement of existing septal defect
Atrial septostomy
Balloon septostomy
creation of septal defect in heart
Blalock–Hanlon procedure
shunt from heart chamber to blood vessel
atrium to pulmonary artery
Fontan procedure
left ventricle to aorta
Rastelli procedure
right ventricle to pulmonary artery
Sano shunt
compound procedures
for transposition of great vessels
Arterial switch operation
Mustard procedure
Senning procedure
for univentricular defect
Norwood procedure
Kawashima procedure
shunt from blood vessel to blood vessel
systemic circulation to pulmonary artery shunt
Blalock–Taussig shunt
SVC to the right PA
Glenn procedure
Cardiac vessels
CHD
Angioplasty
Bypass/Coronary artery bypass
MIDCAB
Off-pump CAB
TECAB
Coronary stent
Bare-metal stent
Drug-eluting stent
Bentall procedure
Valve-sparing aortic root replacement
Other
Pericardium
Pericardiocentesis
Pericardial window
Pericardiectomy
Myocardium
Cardiomyoplasty
Dor procedure
Septal myectomy
Ventricular reduction
Alcohol septal ablation
Conduction system
Maze procedure
Cox maze and minimaze
Catheter ablation
Cryoablation
Radiofrequency ablation
Pacemaker insertion
Left atrial appendage occlusion
Cardiotomy
Heart transplantation
Diagnostic
tests and
procedures
Electrophysiology
Electrocardiography
Vectorcardiography
Holter monitor
Implantable loop recorder
Cardiac stress test
Bruce protocol
Electrophysiology study
Cardiac imaging
Angiocardiography
Echocardiography
TTE
TEE
Myocardial perfusion imaging
Cardiovascular MRI
Ventriculography
Radionuclide ventriculography
Cardiac catheterization/Coronary catheterization
Cardiac CT
Cardiac PET
sound
Phonocardiogram
Function tests
Impedance cardiography
Ballistocardiography
Cardiotocography
Pacing
Cardioversion
Transcutaneous pacing
show
v
t
e
Medical test: Electrodiagnosis
Heart
Electrocardiography
Vectorcardiography
Magnetocardiography
Central nervous system
Electroencephalography (Intracranial EEG, stereoelectroencephalography)
Magnetoencephalography
Peripheral nervous system
Electromyography (Facial electromyography)
Nerve conduction study
Ears
Electrocochleography
Eyes
Electronystagmography
Electrooculography
Electroretinography
Digestive system
Electrogastrogram
Magnetogastrography
show
v
t
e
Physiology of the cardiovascular system
Heart
Cardiac output
Cardiac cycle
Cardiac output
Heart rate
Stroke volume
Stroke volume
End-diastolic volume
End-systolic volume
Afterload
Preload
Frank–Starling law
Cardiac function curve
Venous return curve
Wiggers diagram
Pressure volume diagram
Ultrasound
Fractional shortening = (End-diastolic dimension
End-systolic dimension) / End-diastolic dimension
Aortic valve area calculation
Ejection fraction
Cardiac index
Heart rate
Cardiac pacemaker
Chronotropic (Heart rate)
Dromotropic (Conduction velocity)
Inotropic (Contractility)
Bathmotropic (Excitability)
Lusitropic (Relaxation)
Conduction
Conduction system
Cardiac electrophysiology
Action potential
cardiac
atrial
ventricular
Effective refractory period
Pacemaker potential
Electrocardiography
P wave
PR interval
QRS complex
QT interval
ST segment
T wave
U wave
Hexaxial reference system
Chamber pressure
Central venous
Right
atrial
ventricular
pulmonary artery
wedge
Left
atrial
ventricular
Aortic
Other
Ventricular remodeling
Vascular system/
Hemodynamics
Blood flow
Compliance
Vascular resistance
Pulse
Perfusion
Blood pressure
Pulse pressure
Systolic
Diastolic
Mean arterial pressure
Jugular venous pressure
Portal venous pressure
Regulation of BP
Baroreflex
Kinin–kallikrein system
Renin–angiotensin system
Vasoconstrictors
Vasodilators
Autoregulation
Myogenic mechanism
Tubuloglomerular feedback
Cerebral autoregulation
Paraganglia
Aortic body
Carotid body
Glomus cell
Authority control
GND: 4014280-2
NDL: 00571014
Retrieved from "https://en.wikipedia.org/w/index.php?title=Electrocardiography&oldid=864528834"
Categories:
Cardiac electrophysiology
Cardiac procedures
Electrodiagnosis
Electrophysiology
Mathematics in medicine
Medical tests
Dutch inventions
Hidden categories:
Articles containing German-language text
Articles containing Greek-language text
CS1 maint: Multiple names: authors list
Use dmy dates from May 2014
All articles with unsourced statements
Articles with unsourced statements from April 2017
Articles with unsourced statements from August 2015
All articles lacking reliable references
Articles lacking reliable references from August 2015
Wikipedia articles with GND identifiers
Wikipedia articles with NDL identifiers
Navigation menu
Personal tools
Not logged in
Talk
Contributions
Create account
Log in
Namespaces
Article
Talk
Variants
Views
Read
Edit
View history
More
Search
Navigation
Main page
Contents
Featured content
Current events
Random article
Donate to Wikipedia
Wikipedia store
Interaction
Help
About Wikipedia
Community portal
Recent changes
Contact page
Tools
What links here
Related changes
Upload file
Special pages
Permanent link
Page information
Wikidata item
Cite this page
Print/export
Create a book
Download as PDF
Printable version
In other projects
Wikimedia Commons
Languages
Afrikaans
العربية
Asturianu
Azərbaycanca
বাংলা
Bân-lâm-gú
Беларуская
Беларуская (тарашкевіца)
Български
Bosanski
Brezhoneg
Català
Čeština
Cymraeg
Dansk
Deutsch
Eesti
Ελληνικά
Español
Esperanto
Euskara
فارسی
Français
Gaeilge
Galego
贛語
한국어
हिन्दी
Hrvatski
Bahasa Indonesia
Italiano
עברית
ಕನ್ನಡ
Қазақша
Kurdî
Кыргызча
Latina
Latviešu
Lëtzebuergesch
Lietuvių
Magyar
Македонски
മലയാളം
मराठी
مصرى
မြန်မာဘာသာ
Nederlands
日本語
Norsk
Norsk nynorsk
Oʻzbekcha/ўзбекча
پښتو
Patois
Polski
Português
Română
Русский
Shqip
Simple English
Slovenčina
Slovenščina
Српски / srpski
Srpskohrvatski / српскохрватски
Suomi
Svenska
தமிழ்
ไทย
Türkçe
Українська
Tiếng Việt
Winaray
粵語
中文
64 more
Edit links
This page was last edited on 17 October 2018, at 20:15 (UTC).
Text is available under the Creative Commons Attribution-ShareAlike License; additional terms may apply. By using this site, you agree to the Terms of Use and Privacy Policy. Wikipedia® is a registered trademark of the Wikimedia Foundation, Inc., a non-profit organization.
Privacy policy
About Wikipedia
Disclaimers
Contact Wikipedia
Developers
Cookie statement
Mobile view
Enable previews

key: html

type: local

value: div

attrs: id class

values: mw-page-base noprint

values: mw-head-base noprint

value: a

attrs: id class lang

values: firstHeading firstHeading en

values: siteSub noprint

attrs: class href

values: mw-jump-link #mw-head

values: mw-jump-link #p-search

attrs: href class title

values:
/wiki/ECG_(disambiguation)
mw-disambig
ECG (disambiguation)

values:
/wiki/EKG_(disambiguation)
mw-disambig
EKG (disambiguation)

attrs: role class

values:
note
hatnote navigation-not-searchable

values:
/wiki/Electrography_(disambiguation)
mw-disambig
Electrography (disambiguation)

attrs: href title

value: p

value: th

attrs: colspan style

value: tr

value: img

attrs: alt src width height srcset data-file-width data-file-height

attrs: href class

value: td

values:
/wiki/Electrical_potential
mw-redirect
Electrical potential

values:
/wiki/Non-invasive_(medical)
mw-redirect
Non-invasive (medical)

values:
/wiki/Pacemaker_cells
mw-redirect
Pacemaker cells

values:
/wiki/Heart_chambers
mw-redirect
Heart chambers

values:
/wiki/Artificial_pacemaker
mw-redirect
Artificial pacemaker

values: checkbox button toctogglecheckbox toctogglecheckbox display:none

attrs: alt src width height class srcset data-file-width data-file-height

values:
/wiki/Cardiac_murmur
mw-redirect
Cardiac murmur

values:
/wiki/Cardiac_dysrhythmia
mw-redirect
Cardiac dysrhythmia

values:
/wiki/Sports_physical
mw-redirect
Sports physical

values:
/wiki/Sudden_cardiac_death
mw-redirect
Sudden cardiac death

values:
/wiki/General_anesthesia
mw-redirect
General anesthesia

values: United States Preventive Services Task Force

values:
/wiki/Coronary_heart_disease
mw-redirect
Coronary heart disease

values:
/wiki/Electrocardiogram_technician
mw-redirect
Electrocardiogram technician

values:
/wiki/Cardiac_surgeons
mw-redirect
Cardiac surgeons

values:
/wiki/Voltage_difference
mw-redirect
Voltage difference

values:
/wiki/Analog_electronics
mw-redirect
Analog electronics

values:
/wiki/Mains_power
mw-redirect
Mains power

values:
/wiki/Electrical_potential_difference
mw-redirect
Electrical potential difference

values: This claim needs references to reliable sources. (April 2017)

values:
noprint Inline-Template Template-Fact
white-space:nowrap;

values:
/wiki/Mid-clavicular_line
mw-redirect
Mid-clavicular line

values:
/wiki/Anterior_axillary_line
mw-redirect
Anterior axillary line

values:
/wiki/Midaxillary_line
mw-redirect
Midaxillary line

values:
/wiki/Electrically_conductive
mw-redirect
Electrically conductive

values: vertical-align: -1.838ex; width:25.895ex; height:5.176ex;

values: vertical-align: -0.505ex; width:13.944ex; height:2.343ex;

values: vertical-align: -0.505ex; width:14.955ex; height:2.343ex;

values: vertical-align: -0.505ex; width:15.946ex; height:2.343ex;

values: vertical-align: -1.838ex; width:43.93ex; height:5.176ex;

values: aVR = RA - \frac{1}{2} (LA + LL) = \frac 32 (RA - V_W)

values: vertical-align: -1.838ex; width:43.568ex; height:5.176ex;

values: aVL = LA - \frac{1}{2} (RA + LL) = \frac 32 (LA - V_W)

values: vertical-align: -1.838ex; width:43.566ex; height:5.176ex;

values: aVF = LL - \frac{1}{2} (RA + LA) = \frac 32 (LL - V_W)

values:
/wiki/Left_atrium
mw-redirect
Left atrium

values:
/wiki/Wolff-Parkinson-White_syndrome
mw-redirect
Wolff-Parkinson-White syndrome

values:
/wiki/Cardiac_arrhythmia#Re-entry
mw-redirect
Cardiac arrhythmia

values: Edit section: Lead locations on an ECG report

values:
/wiki/Diaphragmatic_surface_of_heart
mw-redirect
Diaphragmatic surface of heart

values:
/wiki/Lateral_(anatomy)
mw-redirect
Lateral (anatomy)

values:
/wiki/Sternocostal_surface_of_heart
mw-redirect
Sternocostal surface of heart

values: Electrical conduction system of the heart

values:
/wiki/Population_studies
mw-redirect
Population studies

values:
/wiki/Font_weight
mw-redirect
Font weight

values: Measuring time and voltage with ECG graph paper

values:
/wiki/Normal_sinus_rhythm
mw-redirect
Normal sinus rhythm

values:
/wiki/Cardiac_arrhythmia
mw-redirect
Cardiac arrhythmia

values:
/wiki/Atrium_(anatomy)
mw-redirect
Atrium (anatomy)

values:
/wiki/Wolf-Parkinson-White_syndrome
mw-redirect
Wolf-Parkinson-White syndrome

values:
/wiki/Osborn_wave
mw-redirect
Osborn wave

values:
/wiki/Corrected_QT_interval
mw-redirect
Corrected QT interval

values: Electrocardiography in myocardial infarction

values:
/wiki/Coronary_artery
mw-redirect
Coronary artery

values: Anterior interventricular branch of left coronary artery

values: Circumflex branch of left coronary artery

values: Material near this tag needs references to reliable medical sources. (August 2015)

values: Wikipedia:Identifying reliable sources (medicine)

values: Material near this tag may rely on an unreliable or less reliable medical source. (August 2015)

values:
/wiki/Diagnostic_tests
mw-redirect
Diagnostic tests

values:
/wiki/Sinus_arrhythmia
mw-redirect
Sinus arrhythmia

values:
/wiki/Bradycardia-tachycardia_syndrome
mw-redirect
Bradycardia-tachycardia syndrome

values:
/wiki/Ectopic_atrial_tachycardia
mw-redirect
Ectopic atrial tachycardia

values:
/wiki/Paroxysmal_atrial_tachycardia
mw-redirect
Paroxysmal atrial tachycardia

values: Sinoatrial nodal reentrant tachycardia (page does not exist)

values:
/wiki/Wide_complex_tachycardia
mw-redirect
Wide complex tachycardia

values:
/wiki/First-degree_AV_block
mw-redirect
First-degree AV block

values:
/wiki/Second-degree_AV_block
mw-redirect
Second-degree AV block

values:
/wiki/Third-degree_AV_block
mw-redirect
Third-degree AV block

values: Incomplete right bundle branch block (page does not exist)

values: Incomplete left bundle branch block (page does not exist)

values: Right atrial abnormality (page does not exist)

values: Left atrial abnormality (page does not exist)

values:
/wiki/Digoxin_poisoning
mw-redirect
Digoxin poisoning

values:
/wiki/High_Frequency_QRS
mw-redirect
High Frequency QRS

values: Non-Q wave myocardial infarction (page does not exist)

values:
/wiki/Right_ventricular_strain
mw-redirect
Right ventricular strain

values:
/wiki/St_Mary%27s_Hospital_(London)
mw-redirect
St Mary's Hospital (London)

values:
/wiki/Nobel_Prize_in_Medicine
mw-redirect
Nobel Prize in Medicine

values:
/wiki/Apple_Watch_Series_4
mw-redirect
Apple Watch Series 4

values:
reflist mw-references-wrap
list-style-type: lower-alpha;

values:
nofollow
http://www.amamanualofstyle.com/
external text

values:
nofollow
https://myheart.net/articles/nstemi/
external text

values:
/wiki/PubMed_Identifier
mw-redirect
PubMed Identifier

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/22847227

values:
/wiki/ABIM_Foundation
mw-redirect
ABIM Foundation

values:
nofollow
external text
//doi.org/10.1001%2Fjama.2018.6848

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/29896632

values:
nofollow
external text
//www.worldcat.org/issn/0166-9842

values:
nofollow
external text
//doi.org/10.1152%2Fadvan.00029.2012

values:
nofollow
external text
//www.worldcat.org/issn/1043-4046

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/23209014

values:
nofollow
external text
//doi.org/10.3176%2Feng.2008.2.05

values:
nofollow
external text
//doi.org/10.1002%2Fclc.4960171004

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/8001299

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/11093425

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/10987628

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/25464416

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/17056954

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/17322457

values:
nofollow
external text
//doi.org/10.1186%2F1752-1947-1-72

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/17760955

values:
nofollow
external text
//doi.org/10.1016%2Fj.ahj.2014.02.007

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/24766979

values: Category:CS1 maint: Multiple names: authors list

values:
nofollow
external text
//doi.org/10.1056%2FNEJMc0804737

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/18987380

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/16991463

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/18612490

values:
nofollow
external text
//doi.org/10.1161%2F01.CIR.98.18.1937

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/9799216

values:
nofollow
external text
//doi.org/10.1056%2FNEJM198608143150721

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/3526152

values: -moz-column-width: 30em; -webkit-column-width: 30em; column-width: 30em; list-style-type: decimal;

values:
nofollow
external text
http://en.ecgpedia.org/wiki/Main_Page

values:
nofollow
external text
http://en.ecgpedia.org/wiki/ECG_course

values:
nofollow
external text
http://ekg.academy

values:
nofollow
external text
https://ecg.utah.edu/lesson/1

values: International Federation for Emergency Medicine

values: International Conference on Emergency Medicine

values: American College of Emergency Physicians

values: Australasian College for Emergency Medicine

values: Canadian Association of Emergency Physicians

values: Care of the Critically Ill Surgical Patient

values:
/wiki/Pediatric_Advanced_Life_Support
mw-redirect
Pediatric Advanced Life Support

attrs: alt src title width height srcset data-file-width data-file-height

values: nowraplinks hlist collapsible autocollapse navbox-inner mw-collapsible mw-made-collapsible mw-collapsed

values:
/wiki/Human_heart
mw-redirect
Human heart

values:
/wiki/Mitral_valvuloplasty
mw-redirect
Mitral valvuloplasty

values:
/wiki/Left_ventricle
mw-redirect
Left ventricle

values:
/wiki/Right_ventricle
mw-redirect
Right ventricle

values:
/wiki/Transposition_of_great_vessels
mw-redirect
Transposition of great vessels

values:
/wiki/Systemic_circulation
mw-redirect
Systemic circulation

values:
/wiki/Coronary_disease
mw-redirect
Coronary disease

values: Minimally invasive direct coronary artery bypass surgery

values: Totally endoscopic coronary artery bypass surgery

values: Cardiology diagnostic tests and procedures

values: nowraplinks collapsible autocollapse navbox-inner mw-collapsible mw-made-collapsible mw-collapsed

values:
/wiki/Intracranial_EEG
mw-redirect
Intracranial EEG

values:
/wiki/Pressure_volume_diagram
mw-redirect
Pressure volume diagram

values:
/wiki/End-diastolic_dimension
mw-redirect
End-diastolic dimension

values:
/wiki/End-systolic_dimension
mw-redirect
End-systolic dimension

values:
/wiki/Right_ventricular_pressure
mw-redirect
Right ventricular pressure

values:
/wiki/Pulmonary_artery_pressure
mw-redirect
Pulmonary artery pressure

values:
/wiki/Left_ventricular_pressure
mw-redirect
Left ventricular pressure

values:
nofollow
external text
https://d-nb.info/gnd/4014280-2

values: Category:Articles containing German-language text

values: Category:Articles containing Greek-language text

values: Category:All articles with unsourced statements

values: Category:Articles with unsourced statements from April 2017

values: Category:Articles with unsourced statements from August 2015

values: Category:All articles lacking reliable references

values: Category:Articles lacking reliable references from August 2015

values: Category:Wikipedia articles with GND identifiers

values: Category:Wikipedia articles with NDL identifiers

values: Discussion about edits from this IP address [ctrl-alt-n]

values: A list of edits made from this IP address [ctrl-alt-y]

values: You are encouraged to create an account and log in; however, it is not mandatory

values: You're encouraged to log in; however, it's not mandatory. [ctrl-alt-o]

values:
/wiki/Electrocardiography
View the content page [ctrl-alt-c]
c

values: Discussion about the content page [ctrl-alt-t]

values: p-namespaces navigation vectorTabs p-namespaces-label

values:
p-variants
navigation
vectorMenu emptyPortlet
p-variants-label

values: Past revisions of this page [ctrl-alt-h]

values: p-views navigation vectorTabs p-views-label

values:
p-cactions
navigation
vectorMenu emptyPortlet
p-cactions-label

values:
mw-wiki-logo
/wiki/Main_Page
Visit the main page

values:
/wiki/Main_Page
Visit the main page [ctrl-alt-z]
z

values: Featured content – the best of Wikipedia

values: Find background information on current events

values:
/wiki/Special:Random
Load a random article [ctrl-alt-x]
x

values: portal navigation p-navigation p-navigation-label

values: Guidance on how to use and edit Wikipedia

values: About the project, what you can do, where to find things

values: A list of recent changes in the wiki [ctrl-alt-r]

values: portal navigation p-interaction p-interaction-label

values: List of all English Wikipedia pages containing links to this page [ctrl-alt-j]

values: Recent changes in pages linked from this page [ctrl-alt-k]

values:
/wiki/Wikipedia:File_Upload_Wizard
Upload files [ctrl-alt-u]
u

values: A list of all special pages [ctrl-alt-q]

values: Permanent link to this revision of the page

values: Link to connected data repository item [ctrl-alt-g]

values: Printable version of this page [ctrl-alt-p]

values: portal navigation p-wikibase-otherprojects p-wikibase-otherprojects-label

values:
interlanguage-link interwiki-af
display: none;

values: تخطيط كهربائية القلب – Arabic

values:
interlanguage-link interwiki-ast
display: none;

values:
interlanguage-link interwiki-az
display: none;

values: ইলেকট্রোকার্ডিওগ্রাফি – Bangla

values:
interlanguage-link interwiki-bn
display: none;

values:
interlanguage-link interwiki-zh-min-nan
display: none;

values: Электракардыяграфія – Belarusian

values:
interlanguage-link interwiki-be
display: none;

values: Электракардыяграфія – Belarusian (Taraškievica orthography)

values:
interlanguage-link interwiki-be-x-old
display: none;

values: Електрокардиограма – Bulgarian

values:
interlanguage-link interwiki-bg
display: none;

values:
interlanguage-link interwiki-bs
display: none;

values:
interlanguage-link interwiki-br
display: none;

values:
interlanguage-link interwiki-ca
display: none;

values:
interlanguage-link interwiki-cs
display: none;

values:
interlanguage-link interwiki-cy
display: none;

values:
interlanguage-link interwiki-da
display: none;

values:
interlanguage-link interwiki-et
display: none;

values: Ηλεκτροκαρδιογράφημα – Greek

values:
interlanguage-link interwiki-eo
display: none;

values:
interlanguage-link interwiki-eu
display: none;

values: نوار قلب – Persian

values:
interlanguage-link interwiki-fa
display: none;

values:
interlanguage-link interwiki-fr
display: none;

values:
interlanguage-link interwiki-ga
display: none;

values:
interlanguage-link interwiki-gl
display: none;

values:
interlanguage-link interwiki-gan
display: none;

values:
interlanguage-link interwiki-ko
display: none;

values: इलैक्ट्रोकार्डियोग्राफी – Hindi

values:
interlanguage-link interwiki-hr
display: none;

values: interlanguage-link interwiki-it badge-Q17437796 badge-featuredarticle

values: אלקטרוקרדיוגרם – Hebrew

values: ವಿದ್ಯುತ್ ಹೃಲ್ಲೇಖನ – Kannada

values:
interlanguage-link interwiki-kn
display: none;

values: Электрокардиография – Kazakh

values:
interlanguage-link interwiki-kk
display: none;

values:
https://ku.wikipedia.org/wiki/EKG
EKG – Kurdish
ku
ku
interlanguage-link-target

values:
interlanguage-link interwiki-ku
display: none;

values: Электр-кардиограмма (ЭКГ) – Kyrgyz

values:
interlanguage-link interwiki-ky
display: none;

values:
interlanguage-link interwiki-la
display: none;

values:
interlanguage-link interwiki-lv
display: none;

values: Електрокардиографија – Macedonian

values:
interlanguage-link interwiki-mk
display: none;

values:
interlanguage-link interwiki-ml
display: none;

values: छत्स्पंद विद्युत लेखा – Marathi

values:
interlanguage-link interwiki-mr
display: none;

values:
interlanguage-link interwiki-arz
display: none;

values: အီလက်ထရို ကာဒီယိုဂရပ်ဖီ – Burmese

values:
interlanguage-link interwiki-my
display: none;

values:
interlanguage-link interwiki-ja
display: none;

values:
interlanguage-link interwiki-nn
display: none;

values:
interlanguage-link interwiki-uz
display: none;

values: زړه گراف – Pashto

values:
interlanguage-link interwiki-ps
display: none;

values: Ilekchokaadiagrafi – Jamaican Creole English

values:
interlanguage-link interwiki-jam
display: none;

values:
interlanguage-link interwiki-pt
display: none;

values:
interlanguage-link interwiki-ro
display: none;

values: Электрокардиография – Russian

values:
interlanguage-link interwiki-sq
display: none;

values:
interlanguage-link interwiki-sk
display: none;

values:
interlanguage-link interwiki-sl
display: none;

values:
interlanguage-link interwiki-sr
display: none;

values: இதய துடிப்பலைஅளவி – Tamil

values:
interlanguage-link interwiki-ta
display: none;

values: การบันทึกคลื่นไฟฟ้าหัวใจ – Thai

values:
interlanguage-link interwiki-th
display: none;

values: Електрокардіографія – Ukrainian

values:
interlanguage-link interwiki-vi
display: none;

values:
interlanguage-link interwiki-war
display: none;

values:
interlanguage-link interwiki-zh-yue
display: none;

values: /static/images/wikimedia-button-1.5x.png 1.5x, /static/images/wikimedia-button-2x.png 2x

values: /static/images/poweredby_mediawiki_132x47.png 1.5x, /static/images/poweredby_mediawiki_176x62.png 2x

values:
suggestions
display: none; font-size: 13px;

values: mediawiki ltr sitedir-ltr mw-hide-empty-elt ns-0 ns-subject page-Electrocardiography rootpage-Electrocardiography skin-vector action-view