text:
Artificial heart
From Wikipedia, the free encyclopedia
Jump to: navigation, search
This article is about the mechanical device. For the Jonathan Coulton album, see Artificial Heart (album).
The SynCardia temporary Total Artificial Heart
An artificial heart displayed at the London Science Museum
An artificial heart is a device that replaces the heart. Artificial hearts are typically used to bridge the time to heart transplantation, or to permanently replace the heart in case heart transplantation is impossible. Although other similar inventions preceded it from the late 1940s, the first artificial heart to be successfully implanted in a human was the Jarvik-7 in 1982, designed by a team including Willem Johan Kolff and Robert Jarvik.
An artificial (VLAD) is distinct from a ventricular assist device designed to support a failing heart. It is also distinct from a cardiopulmonary bypass machine, which is an external device used to provide the functions of both the heart and lungs and are used only for a few hours at a time, most commonly during cardiac surgery.
Contents
[hide]
1 History
1.1 Origins
1.2 Early development
1.3 Early designs of total artificial hearts
1.4 First clinical implantation of a total artificial heart
1.5 First clinical applications of a permanent pneumatic total artificial heart
1.6 First clinical application of an intrathoracic pump
1.7 First clinical application of a paracorporeal pump
1.8 First VAD patient with FDA approved hospital discharge
2 Total artificial heart prototypes
2.1 POLVAD
2.2 Phoenix-7
2.3 Abiomed AbioCor
2.4 SynCardia
2.5 MagScrew
2.6 Abiomed AbioCor II
2.7 Carmat bioprosthetic heart
2.8 Frazier-Cohn
2.9 Soft Artificial Heart
3 Others
3.1 Hybrid assistive devices
4 See also
5 References
6 External links
History[edit]
Origins[edit]
A synthetic replacement for the heart remains a long-sought "holy grail" of modern medicine. The obvious benefit of a functional artificial heart would be to lower the need for heart transplants, because the demand for organs always greatly exceeds supply (rather necessary for transplant are normally unfit for transfer).
Although the heart is conceptually a pump, it embodies subtleties that defy straightforward emulation with synthetic materials and power supplies. Consequences of these issues include severe foreign-body rejection and external batteries that limit mobility. These complications limited the lifespan of early human recipients to hours or days.
Early development[edit]
The first artificial heart was made by the Soviet scientist Vladimir Demikhov in 1937. It was transplanted to a dog.
On July 2, 1952, 41-year-old Henry Opitek, suffering from shortness of breath, made medical history at Harper University Hospital at Wayne State University in Michigan. The Dodrill-GMR heart machine, considered to be the first operational mechanical heart, was successfully used while performing heart surgery.[1][2]
Forest Dewey Dodrill, working closely with Matthew Dudley, used the machine in 1952 to bypass Henry Opitek's left ventricle for 50 minutes while he opened the patient's left atrium and worked to repair the mitral valve. In Dodrill's post-operative report, he notes, "To our knowledge, this is the first instance of survival of a patient when a mechanicaly heart mechanism was used to take over the complete body function of maintaining the blood supply of the body while the heart was open and operated on."[3]
A heart–lung machine was first used in 1953 during a successful open heart surgery. John Heysham Gibbon, the inventor of the machine, performed the operation and developed the heart–lung substitute himself.
Following these advances, scientific interest for the development of a solution for heart disease developed in numerous research groups worldwide.
Early designs of total artificial hearts[edit]
In 1949, a precursor to the modern artificial heart pump was built by doctors William Sewell and William Glenn of the Yale School of Medicine using an Erector Set, assorted odds and ends, and dime-store toys. The external pump successfully bypassed the heart of a dog for more than an hour.[4]
Paul Winchell invented an artificial heart with the assistance of Henry Heimlich (the inventor of the Heimlich Maneuver) and held the first patent for such a device. The University of Utah developed a similar apparatus around the same time, but when they tried to patent it, Winchell's heart was cited as prior art. The university requested that Winchell donate the heart to the University of Utah, which he did. There is some debate as to how much of Winchell's design Robert Jarvik used in creating Jarvik's artificial heart. Heimlich states, "I saw the heart, I saw the patent and I saw the letters. The basic principle used in Winchell's heart and Jarvik's heart is exactly the same.[5]" Jarvik denies that any of Winchell's design elements were incorporated into the device he fabricated for humans which was successfully implanted into Barney Clark in 1982.
On December 12, 1957, Willem Johan Kolff, the world's most prolific inventor of artificial organs, implanted an artificial heart into a dog at Cleveland Clinic. The dog lived for 90 minutes.
In 1958, Domingo Liotta initiated the studies of TAH replacement at Lyon, France, and in 1959–60 at the National University of Córdoba, Argentina. He presented his work at the meeting of the American Society for Artificial Internal Organs held in Atlantic City in March 1961. At that meeting, Liotta described the implantation of three types of orthotopic (inside the pericardial sac) TAHs in dogs, each of which used a different source of external energy: an implantable electric motor, an implantable rotating pump with an external electric motor, and a pneumatic pump.[6][7]
In 1964, the National Institutes of Health started the Artificial Heart Program, with the goal of putting a man-made heart into a human by the end of the decade.[8] The purpose of the program was to develop an implantable artificial heart, including the power source, to replace a failing heart.[9]
In February 1966, Adrian Kantrowitz rose to international prominence when he performed the world's first permanent implantation of a partial mechanical heart (left ventricular assist device) at Maimonides Medical Center.[10]
In 1967, Kolff left Cleveland Clinic to start the Division of Artificial Organs at the University of Utah and pursue his work on the artificial heart.
In 1973, a calf named Tony survived for 30 days on an early Kolff heart.
In 1975, a bull named Burk survived 90 days on the artificial heart.
In 1976, a calf named Abebe lived for 184 days on the Jarvik 5 artificial heart.
In 1981, a calf named Alfred Lord Tennyson lived for 268 days on the Jarvik 5.
Over the years, more than 200 physicians, engineers, students and faculty developed, tested and improved Kolff's artificial heart. To help manage his many endeavors, Kolff assigned project managers. Each project was named after its manager. Graduate student Robert Jarvik was the project manager for the artificial heart, which was subsequently renamed the Jarvik 7.
In 1981, William DeVries submitted a request to the FDA for permission to implant the Jarvik 7 into a human being. On December 2, 1982, Kolff implanted the Jarvik 7 artificial heart into Barney Clark, a dentist from Seattle who was suffering from severe congestive heart failure. Clark lived for 112 days tethered to an external pneumatic compressor, a device weighing some 400 pounds (180 kg), but during that time he suffered prolonged periods of confusion and a number of instances of bleeding, and asked several times to be allowed to die.[11]
First clinical implantation of a total artificial heart[edit]
On April 4, 1969, Domingo Liotta and Denton A. Cooley replaced a dying man's heart with a mechanical heart inside the chest at The Texas Heart Institute in Houston as a bridge for a transplant. The man woke up and began to recover. After 64 hours, the pneumatic-powered artificial heart was removed and replaced by a donor heart. However thirty-two hours after transplantation, the man died of what was later proved to be an acute pulmonary infection, extended to both lungs, caused by fungi, most likely caused by an immunosuppressive drug complication.[12]
The original prototype of Liotta-Cooley artificial heart used in this historic operation is prominently displayed in the Smithsonian Institution's National Museum of American History "Treasures of American History" exhibit in Washington, D.C.[13]
First clinical applications of a permanent pneumatic total artificial heart[edit]
The first clinical use of an artificial heart designed for permanent implantation rather than a bridge to transplant occurred in 1982 at the University of Utah. Artificial kidney pioneer Willem Johan Kolff started the Utah artificial organs program in 1967.[14] There, physician-engineer Clifford Kwan-Gett invented two components of an integrated pneumatic artificial heart system: a ventricle with hemispherical diaphragms that did not crush red blood cells (a problem with previous artificial hearts) and an external heart driver that inherently regulated blood flow without needing complex control systems.[15] Independently, Paul Winchell designed and patented a similarly shaped ventricle and donated the patent to the Utah program.[16] Throughout the 1970s and early 1980s, veterinarian Donald Olsen led a series of calf experiments that refined the artificial heart and its surgical care. During that time, as a student at the University of Utah, Robert Jarvik combined several modifications: an ovoid shape to fit inside the human chest, a more blood-compatible polyurethane developed by biomedical engineer Donald Lyman, and a fabrication method by Kwan-Gett that made the inside of the ventricles smooth and seamless to reduce dangerous stroke-causing blood clots.[17] On December 2, 1982, William DeVries implanted the artificial heart into retired dentist Barney Bailey Clark (born January 21, 1921), who survived 112 days with the device, dying on March 23, 1983. Bill Schroeder became the second recipient and lived for a record 620 days.
Contrary to popular belief and erroneous articles in several periodicals, the Jarvik heart was not banned for permanent use. Today, the modern version of the Jarvik 7 is known as the SynCardia temporary Total Artificial Heart. It has been implanted in more than 1,350 people as a bridge to transplantation.
In the mid-1980s, artificial hearts were powered by dishwasher-sized pneumatic power sources whose lineage went back to Alfa Laval milking machines. Moreover, two sizable catheters had to cross the body wall to carry the pneumatic pulses to the implanted heart, greatly increasing the risk of infection. To speed development of a new generation of technologies, the National Heart, Lung, and Blood Institute opened a competition for implantable electrically powered artificial hearts. Three groups received funding: Cleveland Clinic in Cleveland, Ohio; the College of Medicine of Pennsylvania State University (Penn State Hershey Medical Center) in Hershey, Pennsylvania; and AbioMed, Inc. of Danvers, Massachusetts. Despite considerable progress, the Cleveland program was discontinued after the first five years.
Polymeric trileaflet valves ensure unidirectional blood flow with a low pressure gradient and good longevity. State-of-the-art transcutaneous energy transfer eliminates the need for electric wires crossing the chest wall.
First clinical application of an intrathoracic pump[edit]
On July 19, 1963, E. Stanley Crawford and Domingo Liotta implanted the first clinical Left Ventricular Assist Device (LVAD) at The Methodist Hospital in Houston, Texas, in a patient who had a cardiac arrest after surgery. The patient survived for four days under mechanical support but did not recover from the complications of the cardiac arrest; finally, the pump was discontinued, and the patient died.
First clinical application of a paracorporeal pump[edit]
1966 DeBakey ventricular assist device.[18]
On April 21, 1966, Michael DeBakey and Liotta implanted the first clinical LVAD in a paracorporeal position (where the external pump rests at the side of the patient) at The Methodist Hospital in Houston, in a patient experiencing cardiogenic shock after heart surgery. The patient developed neurological and pulmonary complications and died after few days of LVAD mechanical support. In October 1966, DeBakey and Liotta implanted the paracorporeal Liotta-DeBakey LVAD in a new patient who recovered well and was discharged from the hospital after 10 days of mechanical support, thus constituting the first successful use of an LVAD for postcardiotomy shock.
First VAD patient with FDA approved hospital discharge[edit]
In 1990 Brian Williams was discharged from the University of Pittsburgh Medical Center (UPMC), becoming the first VAD patient to be discharged with Food and Drug Administration (FDA) approval.[19] The patient was supported in part by bioengineers from the University of Pittsburgh's McGowan Institute.[19][20]
Total artificial heart prototypes[edit]
POLVAD[edit]
Since 1991, the Foundation for Cardiac Surgery Development (FRK) in Zabrze, Poland has been working on developing an artificial heart. Nowadays, the Polish system for heart support POLCAS consists of the artificial ventricle POLVAD-MEV and the three controllers POLPDU-401, POLPDU-402 and POLPDU-501. Presented devices are designed to handle only one patient. The control units of the 401 and 402 series may be used only in hospital due to its big size, method of control and type of power supply. The control[21] unit of 501 series is the latest product of FRK. Due to its much smaller size and weight, it is significantly more mobile solution. For this reason, it can be also used during supervised treatment conducted outside the hospital.
Phoenix-7[edit]
In June 1996, a 46-year-old man received a total artificial heart implantation done by Jeng Wei at Cheng-Hsin General Hospital[22] in the Republic of China (Taiwan). This technologically advanced pneumatic Phoenix-7 Total Artificial Heart was manufactured by a Taiwanese dentist Kelvin K. Cheng, a Chinese physician T. M. Kao and colleagues at the Taiwan TAH Research Center in Tainan, Republic of China (Taiwan). With this experimental artificial heart, the patient's BP was maintained at 90-100/40-55 mmHg and cardiac output at 4.2–5.8 L/min.[23] The patient then received the world's first successful combined heart and kidney transplantation after bridging with a total artificial heart.[24]
Abiomed AbioCor[edit]
The first AbioCor to be surgically implanted in a patient was on July 3, 2001. [25] The AbioCor is made of titanium and plastic with a weight of 0,9 kg (two pounds), and its internal battery can be recharged with a transduction device that sends power through the skin.[25] The internal battery lasts for a half-hour, and a wearable external battery pack lasts for four hours.[26] The FDA announced on September 5, 2006, that the AbioCor could be implanted for humanitarian uses after the device had been tested on 15 patients.[27] It is intended for critically ill patients who cannot receive a heart transplant.[27] Some limitations of the current AbioCor are that its size makes it suitable for only about 50% of the male population, and its useful life is only 1–2 years.[28]
SynCardia[edit]
SynCardia is a company based in Tucson, Arizona which currently has two separate models available. It is available in a 70cc and 50cc size. The 70 cc is used for biventricular heart failure in adult men, while the 50cc is for children and women.[29] As good results with the TAH as a bridge to heart transplant accumulated, a trial of the CardioWest TAH (developed from the Jarvik 7 and now marketed as the Syncardia TAH) was initiated in 1993 and completed in 2002.[30] As of 2014, more than 1,250 patients have received SynCardia artificial hearts.[31][32][31] The device requires the use of the Companion 2 hospital driver or the Freedom portable driver to power the heart with pulses of air. The drivers also monitor blood flow for each ventricle.[33]
MagScrew[edit]
Another U.S. team has a prototype called the 2005 MagScrew Total Artificial Heart. Teams in Japan and South Korea are also racing to produce similar devices.[34][35][36][37]
Abiomed AbioCor II[edit]
By combining its valved ventricles with the control technology and roller screw developed at Penn State, AbioMed has designed a smaller, more stable heart, the AbioCor II. This pump, which should be implantable in most men and 50% of women with a life span of up to five years,[28] had animal trials in 2005, and the company hoped to get FDA approval for human use in 2008.[38]
Carmat bioprosthetic heart[edit]
On October 27, 2008, French professor and leading heart transplant specialist Alain F. Carpentier announced that a fully implantable artificial heart would be ready for clinical trial by 2011 and for alternative transplant in 2013. It was developed and would be manufactured by him, biomedical firm CARMAT SA,[39] and venture capital firm Truffle Capital. The prototype used embedded electronic sensors and was made from chemically treated animal tissues, called "biomaterials", or a "pseudo-skin" of biosynthetic, microporous materials.[40]
According to a press-release by Carmat dated December 20, 2013, the first implantation of its artificial heart in a 75-year-old patient was performed on December 18, 2013 by the Georges Pompidou European Hospital team in Paris (France).[41] The patient died 75 days after the operation.[42]
In Carmat's design, two chambers are each divided by a membrane that holds hydraulic fluid on one side. A motorized pump moves hydraulic fluid in and out of the chambers, and that fluid causes the membrane to move; blood flows through the other side of each membrane. The blood-facing side of the membrane is made of tissue obtained from a sac that surrounds a cow's heart, to make the device more biocompatible. The Carmat device also uses valves made from cow heart tissue and has sensors to detect increased pressure within the device. That information is sent to an internal control system that can adjust the flow rate in response to increased demand, such as when a patient is exercising.[43] This distinguishes it from previous designs that maintain a constant flow rate.
The Carmat device, unlike previous designs, is meant to be used in cases of terminal heart failure, instead of being used as a bridge device while the patient awaits a transplant.[44] At 900 grams it weighs nearly three times the typical heart and is targeted primarily towards obese men. It also requires the patient to carry around an additional Li-Ion battery. The projected lifetime of the artificial heart is around 5 years (230 million beats).[45]
Frazier-Cohn[edit]
On 12 March 2011, an experimental artificial heart was implanted in 55-year-old Craig Lewis at The Texas Heart Institute in Houston by Drs. O. H. Frazier and William Cohn. The device is a combination of two modified HeartMate II pumps that is currently undergoing bovine trials.[46]
Frazier and Cohn are on the board of the BiVACOR company that develops an artificial heart.[47][48] BiVACOR has been tested as a replacement for a heart in a sheep.[49][50]
So far, only one person has benefited from Frazier and Cohn's artificial heart. Craig Lewis was suffering from amyloidosis in 2011 when his heart gave out and doctors pronounced that he had only 12 to 24 hours to live. After obtaining permission from his family, Frazier and Cohn replaced his heart with their device. Lewis survived for another 5 weeks after the operation; he eventually succumbed to liver and kidney failure due to his amyloidosis, after which his family asked that his artificial heart be unplugged.[51]
Soft Artificial Heart[edit]
On 10 July 2017, Cohrs and colleagues presented a new concept of a soft total artificial heart in the Journal of Artificial Organs.[52] The heart was developed in the Functionals Materials Laboratory at ETH Zurich.[53] The soft artificial heart (SAH) was created from silicone with the help of 3D printing technology. The SAH is a silicone monoblock. It weighs 390g, has a volume of 679 cm^3 and is operated through pressurized air. "Our goal is to develop an artificial heart that is roughly the same size as the patient’s own one and which imitates the human heart as closely as possible in form and function" says Cohrs in an interview.[54] The SAH fundamentally moves and works like a real heart but currently only beats for 3000 beats.[55]
Others[edit]
A centrifugal pump[56][57] or an axial-flow pump[58][59] can be used as an artificial heart, resulting in the patient being alive without a pulse.
A centrifugal artificial heart which alternately pumps the pulmonary circulation and the systemic circulation, causing a pulse has been described.[60]
Researchers have constructed a heart out of foam. The heart is made out of flexible silicone and works with an external pump to push air and fluids through the heart. It currently cannot be implanted into humans, but it is a promising start for artificial hearts.[61]
Hybrid assistive devices[edit]
Main article: Ventricular assist device
Patients who have some remaining heart function but who can no longer live normally may be candidates for ventricular assist devices (VAD), which do not replace the human heart but complement it by taking up much of the function.
The first Left Ventricular Assist Device (LVAD) system was created by Domingo Liotta at Baylor College of Medicine in Houston in 1962.[62]
Another VAD, the Kantrowitz CardioVad, designed by Adrian Kantrowitz boosts the native heart by taking up over 50% of its function.[63] Additionally, the VAD can help patients on the wait list for a heart transplant. In a young person, this device could delay the need for a transplant by 10–15 years, or even allow the heart to recover, in which case the VAD can be removed.[63] The artificial heart is powered by a battery that needs to be changed several times while still working.
The first heart assist device was approved by the FDA in 1994, and two more received approval in 1998.[64] While the original assist devices emulated the pulsating heart, newer versions, such as the Heartmate II,[65] developed by The Texas Heart Institute of Houston, provide continuous flow. These pumps (which may be centrifugal or axial flow) are smaller and potentially more durable and last longer than the current generation of total heart replacement pumps. Another major advantage of a VAD is that the patient keeps the natural heart, which may still function for temporary back-up support if the mechanical pump were to stop. This may provide enough support to keep the patient alive until a solution to the problem is implemented.
In August 2006, an artificial heart was implanted into a 15-year-old girl at the Stollery Children's Hospital in Edmonton, Alberta. It was intended to act as a temporary fixture until a donor heart could be found. Instead, the artificial heart (called a Berlin Heart) allowed for natural processes to occur and her heart healed on its own. After 146 days, the Berlin Heart was removed, and the girl's heart functioned properly on its own.[66] On December 16, 2011 the Berlin Heart gained U.S. FDA approval. The device has since been successfully implanted in several children including a 4-year-old Honduran girl at Children's Hospital Boston.[67]
Several continuous-flow ventricular assist devices have been approved for use in the European Union, and, as of August 2007, were undergoing clinical trials for FDA approval.
In 2012, a study published in the New England Journal of Medicine compared the Berlin Heart to extracorporeal membrane oxygenation (ECMO) and concluded that "a ventricular assist device available in several sizes for use in children as a bridge to heart transplantation [such as the Berlin Heart] was associated with a significantly higher rate of survival as compared with ECMO."[68] The study's primary author, Charles D. Fraser, Jr., surgeon in chief at Texas Children's Hospital, explained: "With the Berlin Heart, we have a more effective therapy to offer patients earlier in the management of their heart failure. When we sit with parents, we have real data to offer so they can make an informed decision. This is a giant step forward." [69]
Suffering from end-stage heart failure, former Vice President Dick Cheney underwent a procedure in July 2010 to have a VAD implanted at INOVA Fairfax Hospital, in Fairfax Virginia. In 2012, he received a heart transplant at age 71 after 20 months on a waiting list.
See also[edit]
Artificial heart valve
References[edit]
Jump up ^ American Heart Association. The Mechanical Heart celebrates 50 lifesaving years. 22 10 2002. Original link: <http://www.americanheart.org/presenter.jhtml;jsessionid=EFNP3NSFUBXLICQFCXQCDSQ?identifier=3005888>(9 Feb 2008). Still available at [1] and [2].
Jump up ^ "50th Anniversary of First Open Heart Surgery – Newsroom".
Jump up ^ Stephenson Larry W; et al. (2002). "The Michigan Heart: The World's First Successful Open Heart Operation?". Journal of Cardiac Surgery. 17 (3): 238–246. PMID 12489911. doi:10.1111/j.1540-8191.2002.tb01209.x.
Jump up ^ Lavietes, Stuart. William Glenn, 88, Surgeon Who Invented Heart Procedure, The New York Times, March 17, 2003. Accessed May 21, 2009.
Jump up ^ Jarvik-7 schematic
Jump up ^ "Artificial Heart in the chest: Preliminary report". Trans. Amer. Soc. Inter. Organs. 7: 318. 1961.
Jump up ^ "Ablation experimentale et replacement du coeur par un coer artificial intra-thoracique". Lyon Cirurgical. 57: 704. 1961.
Jump up ^ Sandeep Jauhar: The Artificial Heart. New England Journal of Medicine (2004): 542–544.
Jump up ^ Hwang, Ned H. C.; Woo, Savio L.-Y. (31 December 2003). "Frontiers in Biomedical Engineering: Proceedings of the World Congress for Chinese Biomedical Engineers". Springer Science & Business Media – via Google Books.
Jump up ^ "In Memoriam: Adrian Kantrowitz 1918–2008". Texas Heart Institute Journal. 36 (1): 1. PMC 2676518  .
Jump up ^ Barron H. Lerner, MD (December 1, 2007). "The 25th Anniversary of Barney Clark's Artificial Heart". Celebrity Health. HealthDiaries.com. Retrieved 15 November 2010.
Jump up ^ Orthotopic cardiac prosthesis for two-staged cardiac replacement. Am J Cardio 1969; 24:723–730.
Jump up ^ "Treasures of American History", National Museum of American History
Jump up ^ Spare Parts: Organ Replacement in American Society. Renee C. Fox and Judith P. Swazey. New York: Oxford University Press; 1992, pp. 102–104
Jump up ^ Kwan-Gett CS, Van Kampen KR, Kawai J, Eastwood N, Kolff WJ (Dec 1971). "Results of total artificial heart implantation in calves". Journal of Thoracic and Cardiovascular Surgery. 62 (6): 880–889.
Jump up ^ "Winchell's Heart". Time. March 12, 1973. Retrieved April 25, 2010.
Jump up ^ "Kolff".
Jump up ^ "Dr. Denton Cooley and Dr. Michael E. DeBakey: Rock stars of Houston medicine". Houston Chronicle. 3 April 2014. Retrieved 7 March 2015.
^ Jump up to: a b "Revolutionizing Care for Failing Hearts – William G. McGowan Charitable Fund".
Jump up ^ "Pittsburgh’s 25th Anniversary of First VAD Patient to Be Discharged from a Hospital – Regenerative Medicine at the McGowan Institute".
Jump up ^ Fajdek, Bartlomiej; Janiszowski, Krzysztof. "Automatic control system for ventricular assist device". IEEE. doi:10.1109/MMAR.2014.6957472.
Jump up ^ "Cheng-Hsin General Hospital".
Jump up ^ http://www.worldscientific.com/doi/pdf/10.4015/S1016237201000170
Jump up ^ Wei J.; Cheng K. K.; Tung D. Y.; Chang C. Y.; Wan W. M.; Chuang Y. C. (1998). "Successful Use of Phoenix-7 Total Artificial Heart". Transplantation Proceedings. 30: 3403–4. doi:10.1016/s0041-1345(98)01078-1.
^ Jump up to: a b "Patient gets first totally implanted artificial heart". CNN.com. 2001-07-03. Archived from the original on 7 June 2008. Retrieved 2008-07-13.
Jump up ^ "AbioCor FAQs". AbioMed. Archived from the original on 3 July 2008. Retrieved 2008-07-13.
^ Jump up to: a b "FDA Approves First Totally Implanted Permanent Artificial Heart for Humanitarian Uses". FDA.gov. 2006-09-05. Retrieved 2008-07-13.
^ Jump up to: a b "Will We Merge With Machines?". popsci.com. 2005-08-01. Archived from the original on 19 July 2008. Retrieved 2008-07-13.
Jump up ^ "Two Sizes Intended to Treat Most Patients". SynCardia. November 16, 2015.
Jump up ^ Copeland JG, Smith RG, Arabia FA, et al. (2004). "Cardiac replacement with a total artificial heart as a bridge to transplantation". N Engl J Med. 351: 859–67. doi:10.1056/nejmoa040186.
^ Jump up to: a b "New Record of 161 SynCardia Total Artificial Heart Implants Set In 2013". SynCardia. 2014-01-07. Retrieved 2016-06-11.
Jump up ^ Torregrossa G, Morshuis M, Varghese R; et al. (2014). "Results with SynCardia total artificial heart beyond 1 year". ASAIO J. 60: 626–34. CS1 maint: Explicit use of et al. (link)CS1 maint: Multiple names: authors list (link) One patient had the artificial heart for 1,374 days, that is, nearly four years.
Jump up ^ SynCardia drivers
Jump up ^ Weber S, Kamohara K, Klatte RS, Luangphakdy V, Flick C, Chen JF, Casas F, Ootaki Y, Kopcak M, Akiyama M, Hirschman GB, Chapman PA, Donahue A, Wetterau W, Prisco C, Mast R, Sherman C, Fukamachi K, Smith WA. "MagScrew TAH: an update". ASAIO J. 51: xxxvi–xlvi. PMID 16340348. doi:10.1097/01.mat.0000187395.29817.36.
Jump up ^ Helical Flow Total Artificial Heart
Jump up ^ HFTAH
Jump up ^ South Korean TAH
Jump up ^ "14th Artificial Heart Patient Dies: A Newsmaker Interview With Robert Kung, PhD". medscape.com. 2004-11-11. Retrieved 2008-07-13.
Jump up ^ atorsoli, Albertina Torsoli. "A Synthetic Heart That Keeps on Ticking".
Jump up ^ "The Carmat Heart,- The technology behind the prosthesis". Archived from the original on 2013-06-29.
Jump up ^ "First-in-man implantation of CARMAT's bioprosthetic artificial heart". Archived from the original on 2013-12-24.
Jump up ^ "First patient fitted with Carmat artificial heart dies". Reuters. March 3, 2014.
Jump up ^ Rojahn, Susan Young. "Biology and Machine Come Together in a New Artificial Heart".
Jump up ^ "Carmat artificial heart patient in good condition: hospital". Reuters. December 30, 2013.
Jump up ^ Polska, Wirtualna. "Pierwsze wszczepione na stałe sztuczne serce bije już w ludzkiej piersi – TechTrendy.pl".
Jump up ^ Berger, Eric. "New artificial heart 'a leap forward'". Houston Chronicle. Retrieved 23 March 2011.
Jump up ^ "Donor brings inventor of artificial heart closer to Houston". Houston Chronicle. 14 Jan 2013.
Jump up ^ "Living without a pulse: Engineering a better artificial heart". CNN. 4 Dec 2013.
Jump up ^ "BiVACOR beatless artificial heart appeal hopes to raise $5 million". Courier Mail. 7 March 2015.
Jump up ^ "Bionic heart breakthrough: Scientists transplant device into sheep, hope for clinical trials". ABC News. Australian Broadcasting Corporation. 7 March 2015.
Jump up ^ "Walking Dead – The First Human Able to Survive Without a Heart OR a Pulse". Bam Marguera. Archived from the original on 5 April 2015. Retrieved 4 April 2015.
Jump up ^ Cohrs Nicholas H. "A Soft Total Artificial Heart-First Concept Evaluation on a Hybrid Mock Circulation". Artificial Organs. doi:10.1111/aor.12956.
Jump up ^ http://www.fml.ethz.ch/news-and-events/fml-news/2017/07/news-article-featuring-nicholas-cohrs-and-his-soft-artificial-heart.html
Jump up ^ https://www.ethz.ch/en/news-and-events/eth-news/news/2017/07/artificial_heart.html
Jump up ^ https://www.youtube.com/watch?v=YUYNXeHfTdQ
Jump up ^ Black, Rosemary (January 5, 2011). "Former vice president Dick Cheney now has no pulse". Daily News. New York.
Jump up ^ "Pulseless Pumps & Artificial Hearts".
Jump up ^ European-Hospital. "The pulseless life on healthcare in europe".
Jump up ^ Dan Baum: No Pulse: How Doctors Reinvented The Human Heart. 2012-02-29.
Jump up ^ Imachi K, Chinzei T, Abe Y, Mabuchi K, Imanishi K, Yonezawa T, Kouno A, Ono T, Atsumi K, Isoyama T. "A new pulsatile total artificial heart using a single centrifugal pump". ASAIO Trans. 37: M242–3. PMID 1751129.
Jump up ^ "New Artificial Heart is Made of Foam". Popular Science. 1 October 2015.
Jump up ^ Prolonged Assisted circulation after cardiac or aortic surgery. Prolonged partial left ventricular bypass by means of intracorporeal circulation. This paper was finalist in The Young Investigators Award Contest of the American College of Cardiology. Denver, May 1962 Am. J. Cardiol. 1963, 12:399–404
^ Jump up to: a b Mitka Mike (2001). "Midwest Trials of Heart-Assist Device". Journal of the American Medical Association. 286 (21): 2661. doi:10.1001/jama.286.21.2661.
Jump up ^ FDA APPROVES TWO PORTABLE HEART-ASSIST DEVICES at FDA.gov
Jump up ^ An Artificial Heart That Doesn't Beat at TechnologyReview.com
Jump up ^ "Archived copy". Archived from the original on 2007-10-16. Retrieved 2007-08-29. CS1 maint: BOT: original-url status unknown (link) (August 28, 2007), Capital Health, Edmonton (archived from "Archived copy". Archived from the original on 2007-09-27. Retrieved 2007-08-29. the original) on 2007-10-01.
Jump up ^ "Newly approved Berlin Heart helps patients waiting for a transplant – Vector". 30 December 2011.
Jump up ^ Fraser, Charles D.; Jaquiss, Robert D.B.; Rosenthal, David N.; Humpl, Tilman; Canter, Charles E.; Blackstone, Eugene H.; Naftel, David C.; Ichord, Rebecca N.; Bomgaars, Lisa; Tweddell, James S.; Massicotte, M. Patricia; Turrentine, Mark W.; Cohen, Gordon A.; Devaney, Eric J.; Pearce, F. Bennett; Carberry, Kathleen E.; Kroslowitz, Robert; Almond, Christopher S. "Prospective Trial of a Pediatric Ventricular Assist Device". New England Journal of Medicine. 367 (6): 532–541. doi:10.1056/nejmoa1014164.
Jump up ^ "News – Texas Children's Hospital".
George B. Griffenhagen and Calvin H. Hughes. The History of the Mechanical Heart. Smithsonian Report for 1955, (Pub. 4241): 339–356, 1956.
External links[edit]
Artificial heart. on YouTube
Artificial hearts and heart assist devices currently in use
Artificial heart a medical marvel – The first person in Australia to receive a total artificial heart transplant.
How a total artificial heart works physics.org
A Change of Heart by Retro Report
Retrieved from "https://en.wikipedia.org/w/index.php?title=Artificial_heart&oldid=806206249"
Categories:
American inventions
Implants (medicine)
Cardiology
Artificial organs
Hidden categories:
CS1 maint: Explicit use of et al.
CS1 maint: Multiple names: authors list
CS1 maint: BOT: original-url status unknown
Navigation menu
Personal tools
Not logged in
Talk
Contributions
Create account
Log in
Namespaces
Article
Talk
Variants
Views
Read
Edit
View history
More
Search
Navigation
Main page
Contents
Featured content
Current events
Random article
Donate to Wikipedia
Wikipedia store
Interaction
Help
About Wikipedia
Community portal
Recent changes
Contact page
Tools
What links here
Related changes
Upload file
Special pages
Permanent link
Page information
Wikidata item
Cite this page
Print/export
Create a book
Download as PDF
Printable version
In other projects
Wikimedia Commons
Languages
العربية
Azərbaycanca
Català
Deutsch
Eesti
Español
فارسی
Français
Gaeilge
한국어
Հայերեն
Bahasa Indonesia
Italiano
Magyar
Nederlands
日本語
پښتو
Português
Русский
Slovenčina
Suomi
Svenska
தமிழ்
粵語
中文
Edit links
This page was last edited on 20 October 2017, at 12:44.
Text is available under the Creative Commons Attribution-ShareAlike License; additional terms may apply. By using this site, you agree to the Terms of Use and Privacy Policy. Wikipedia® is a registered trademark of the Wikimedia Foundation, Inc., a non-profit organization.
Privacy policy
About Wikipedia
Disclaimers
Contact Wikipedia
Developers
Cookie statement
Mobile view
Enable previews

key: html

type: local

value: div

attrs: id class

values: mw-page-base noprint

values: mw-head-base noprint

value: a

attrs: id class lang

values: firstHeading firstHeading en

values: siteSub noprint

values: jump-to-nav mw-jump

attrs: href title

attrs: role class

values:
note
hatnote navigation-not-searchable

value: img

attrs: alt src width height class srcset data-file-width data-file-height

attrs: href class

attrs: class style

values:
/wiki/London_Science_Museum
mw-redirect
London Science Museum

value: b

values:
/wiki/Human_heart
mw-redirect
Human heart

values: Edit section: Early designs of total artificial hearts

values:
/wiki/Congestive_heart_failure
mw-redirect
Congestive heart failure

values: Edit section: First clinical implantation of a total artificial heart

values: Edit section: First clinical applications of a permanent pneumatic total artificial heart

values: National Heart, Lung, and Blood Institute

values:
/wiki/Penn_State_Hershey_Medical_Center
mw-redirect
Penn State Hershey Medical Center

values: Edit section: First clinical application of an intrathoracic pump

values:
/wiki/The_Methodist_Hospital
mw-redirect
The Methodist Hospital

values: Edit section: First clinical application of a paracorporeal pump

values:
/wiki/Michael_E._DeBakey
mw-redirect
Michael E. DeBakey

values: Edit section: First VAD patient with FDA approved hospital discharge

values: Edit section: Total artificial heart prototypes

values: Phoenix-7 Total Artificial Heart (page does not exist)

values: Taiwan TAH Research Center (page does not exist)

values:
/wiki/Heart_transplant
mw-redirect
Heart transplant

values: Edit section: Carmat bioprosthetic heart

values:
/wiki/Alain_F._Carpentier
mw-redirect
Alain F. Carpentier

values:
/wiki/Systemic_circulation
mw-redirect
Systemic circulation

values:
/wiki/Children%27s_Hospital_Boston
mw-redirect
Children's Hospital Boston

values:
/wiki/PubMed_Identifier
mw-redirect
PubMed Identifier

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/12489911

values:
nofollow
external text
//doi.org/10.1109%2FMMAR.2014.6957472

values:
nofollow
external text
//doi.org/10.1056%2Fnejmoa040186

values: Category:CS1 maint: Explicit use of et al.

values: Category:CS1 maint: Multiple names: authors list

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/16340348

values:
nofollow
external text
//doi.org/10.1111%2Faor.12956

values:
nofollow
external text
//www.ncbi.nlm.nih.gov/pubmed/1751129

values:
nofollow
external text
//doi.org/10.1001%2Fjama.286.21.2661

values: Category:CS1 maint: BOT: original-url status unknown

values:
nofollow
external text
//doi.org/10.1056%2Fnejmoa1014164

values: -moz-column-width: 30em; -webkit-column-width: 30em; column-width: 30em; list-style-type: decimal;

values:
nofollow
http://texasheart.org/Research/Devices/
external text

values: Discussion about edits from this IP address [ctrl-alt-n]

values: A list of edits made from this IP address [ctrl-alt-y]

values: You are encouraged to create an account and log in; however, it is not mandatory

values: You're encouraged to log in; however, it's not mandatory. [ctrl-alt-o]

values:
/wiki/Artificial_heart
View the content page [ctrl-alt-c]
c

values: Discussion about the content page [ctrl-alt-t]

values: p-namespaces navigation vectorTabs p-namespaces-label

values:
p-variants
navigation
vectorMenu emptyPortlet
p-variants-label

values: Past revisions of this page [ctrl-alt-h]

values: p-views navigation vectorTabs p-views-label

values:
p-cactions
navigation
vectorMenu emptyPortlet
p-cactions-label

values:
mw-wiki-logo
/wiki/Main_Page
Visit the main page

values:
/wiki/Main_Page
Visit the main page [ctrl-alt-z]
z

values: Featured content – the best of Wikipedia

values: Find background information on current events

values:
/wiki/Special:Random
Load a random article [ctrl-alt-x]
x

values: portal navigation p-navigation p-navigation-label

values: Guidance on how to use and edit Wikipedia

values: About the project, what you can do, where to find things

values: A list of recent changes in the wiki [ctrl-alt-r]

values: portal navigation p-interaction p-interaction-label

values: List of all English Wikipedia pages containing links to this page [ctrl-alt-j]

values: Recent changes in pages linked from this page [ctrl-alt-k]

values:
/wiki/Wikipedia:File_Upload_Wizard
Upload files [ctrl-alt-u]
u

values: A list of all special pages [ctrl-alt-q]

values: Permanent link to this revision of the page

values: Link to connected data repository item [ctrl-alt-g]

values: Printable version of this page [ctrl-alt-p]

values: portal navigation p-wikibase-otherprojects p-wikibase-otherprojects-label

values:
uls-settings-trigger
Language settings
0
button
true

values: قلب صناعي – Arabic

values:
https://de.wikipedia.org/wiki/Kunstherz
Kunstherz – German
de
de
interlanguage-link-target

values: قلب مصنوعی – Persian

values: Արհեստական Սիրտ – Armenian

values:
https://nl.wikipedia.org/wiki/Kunsthart
Kunsthart – Dutch
nl
nl
interlanguage-link-target

values: مصنوعي زړه – Pashto

values: Искусственное сердце – Russian

values: செயற்கை இதயம் – Tamil

values: /static/images/wikimedia-button-1.5x.png 1.5x, /static/images/wikimedia-button-2x.png 2x

values: /static/images/poweredby_mediawiki_132x47.png 1.5x, /static/images/poweredby_mediawiki_176x62.png 2x

values:
suggestions
display: none; font-size: 13px;

values: mediawiki ltr sitedir-ltr mw-hide-empty-elt ns-0 ns-subject page-Artificial_heart rootpage-Artificial_heart vector-experimental-print-styles vector-nav-directionality skin-vector action-view