text:
Vernier scale
From Wikipedia, the free encyclopedia
Jump to: navigation, search
A close-up of a caliper's measurement scales. Assuming the caliper has no "zero error" (that is, it registers 0.00 mm when fully closed) the image shows a reading of 3.58 mm ± 0.02 mm. This is found by adding 3.00 mm (left red mark) read off from the fixed main (upper) scale to 0.58 mm (right red mark) obtained from the sliding vernier (lower) scale. The main scale reading is determined by the rightmost tick on the main scale that is to the left of the zero tick on the vernier scale. The vernier reading is found by locating the closest aligned lines between the two scales. The 0.02 mm inscription indicates the caliper's precision and is just the width that corresponds to the smallest interval on the vernier scale.
A vernier scale is a visual aid that allows the user to measure more precisely than could be done unaided when reading a uniformly divided straight or circular measurement scale. It is a subsidiary scale that indicates where the measurement lies in between two of the graduations on the main scale. Verniers are common on sextants used in navigation, scientific instruments used to conduct experiments, machinists' (or jewelers') measuring tools (all sorts, but especially calipers and micrometers) used to work materials to fine tolerances, on theodolites used in surveying, and in absolute encoders to measure linear or rotational displacements.
Contents
[hide]
1 History
2 Construction
3 Use
4 Least count
5 Examples
6 Mechanism of a vernier scale
7 Vernier acuity
8 Zero error
9 See also
10 References
11 External links
History[edit]
Mechanical displacement gauges with vernier scales on wall cracks (Moika Palace, Saint Petersburg).
Calipers without a vernier scale originated in ancient China as early as the Qin dynasty (AD 9).[1][2] The secondary scale, which contributed extra precision, was invented in 1631 by French mathematician Pierre Vernier (1580–1637). Its use was described in detail in English in Navigatio Britannica (1750) by mathematician and historian John Barrow.[3] While calipers are the most typical use of Vernier scales today, they were originally developed for angle-measuring instruments such as astronomical quadrants.
In some languages, the Vernier scale is called a nonius, a precursor to the Vernier scale developed by Portuguese mathematician, cosmographer Pedro Nunes (1502–1578), latinised as Petrus Nonius. It was known by this name in English until the end of the 18th century.[4]
The name "vernier" was popularised by the French astronomer Jérôme Lalande (1732–1807) through his Traité d'astronomie (2 vols) (1764).[5]
Construction[edit]
In the following, N is the number of divisions the maker wishes to show at a finer level of measure.
Vernier scale direct vernier
Direct verniers are the most common. The indicating scale is constructed so that when its zero point coincides with the start of the data scale, its graduations are at a slightly smaller spacing than those on the data scale and so none but the last graduation coincide with any graduations on the data scale. N graduations of the indicating scale cover N−1 graduations of the data scale.
Retrograde verniers are found on some devices, including surveying instruments.[6] A retrograde vernier is similar to the direct vernier, except its graduations are at a slightly larger spacing than on the main scale. N graduations of the indicating scale cover N+1 graduations of the data scale. The retrograde vernier also extends backwards along the data scale.
Direct and retrograde verniers are read in the same manner.
Use[edit]
Animation of a caliper measurement using a vernier scale. Click to enlarge
Vernier scale use 0.02 scale measurement is 19.44 mm
The main use of the vernier caliper is to measure the internal and the external diameters of an object. To measure using a vernier scale, the user first reads the finely marked "fixed" scale (i.e. the main scale; see the diagram). This measure is typically between two of the main scale's smallest graduations. The user then reads the vernier scale (see diagram), which measures between the smallest graduations on the fixed scale—providing much greater precision.
Least count[edit]
The difference between the value of one main scale division and the value of one Vernier scale division is known as least count of the Vernier. It is also known as Vernier constant. Let the measure of the smallest main scale reading, that is the distance between two consecutive graduations (also called its pitch) be S and the distance between two consecutive Vernier scale graduations be V such that the length of (n-1) main scale divisions is equal to n Vernier scale divisions. Then,
the length of (n-1) main scale divisions = the length of n vernier scale division
or, (n-1)S = nV
or, nS-S = nV
or, S = nS - nV
or, S/n = (S-V)
or (Pitch)/(Number of Vernier scale divisions) = (Length of one main scale division - Length of one Vernier scale division)
S/n and (S-V) are both equal to the least count of vernier scale, and are also called the vernier constant.
Examples[edit]
On decimal measuring instruments, as in the diagram below, the indicating scale has 10 graduations that cover the same length as 9 on the data scale. Note that the vernier's 10th graduation is omitted.
On an angular measurement instrument, the data scale might be in half-degrees, with an indicator scale that provides 30 1-minute graduations (that span 29 of the half-degree graduations).
Mechanism of a vernier scale[edit]
Vernier scale How a vernier scale works
The vernier scale is constructed so that it is spaced at a constant fraction of the fixed main scale. So for a decimal measuring device each mark on the vernier is spaced nine tenths of those on the main scale. If you put the two scales together with zero points aligned, the first mark on the vernier scale is one tenth short of the first main scale mark, the second two tenths short, and so on up to the ninth mark—which is misaligned by nine tenths. Only when a full ten marks are counted is there alignment, because the tenth mark is ten tenths—a whole main scale unit short, and therefore aligns with the ninth mark on the main scale.
Now if you move the vernier by a small amount, say, one tenth of its fixed main scale, the only pair of marks that come into alignment are the first pair, since these were the only ones originally misaligned by one tenth. If we move it two tenths, the second pair aligns, since these are the only ones originally misaligned by that amount. If we move it five tenths, the fifth pair aligns—and so on. For any movement, only one pair of marks aligns and that pair shows the value between the marks on the fixed scale.
Vernier acuity[edit]
Main article: Vernier acuity
Vernier scales work so well because most people are especially good at detecting which of the lines is aligned and misaligned, and that ability gets better with practice, in fact far exceeding the optical capability of the eye. This ability to detect alignment is called 'Vernier acuity'.[7] Historically, none of the alternative technologies exploited this or any other hyperacuity, giving the Vernier scale an advantage over its competitors.[8]
Zero error[edit]
Zero error is defined as the condition where a measuring instrument registers a reading when there should not be any reading. In case of vernier calipers it occurs when a zero on main scale does not coincide with a zero on vernier scale. The zero error may be of two types i.e. when the scale is towards numbers greater than zero it is positive else negative. The method to use a vernier scale or caliper with zero error is to use the formula: actual reading = main scale + vernier scale − (zero error). Zero error may arise due to knocks that cause the calibration to be thrown off at the 0.00 mm when the jaws are perfectly closed or just touching each other.
When the jaws are closed and if the reading is 0.10mm, the zero error is referred to as +0.10mm. The method to use a vernier scale or caliper with zero error is to use the formula 'actual reading = main scale + vernier scale − (zero error)' thus the actual reading is 19.00 + 0.54 − (0.10) = 19.44 mm
Positive zero error refers to the case when the jaws of the vernier caliper are just closed and the reading is a positive reading away from the actual reading of 0.00mm. If the reading is 0.10mm, the zero error is referred to as +0.10 mm.
When the jaws are closed and if the reading is −0.08mm, the zero error is referred to as +0.08 mm. The method to use a vernier scale or caliper with zero error is to use the formula 'actual reading = main scale + vernier scale + (zero error)' thus the actual reading is 19.00 + 0.36 + (+0.08) = 19.44 mm
Negative zero error refers to the case when the jaws of the vernier caliper are just closed and the reading is a negative reading away from the actual reading of 0.00mm. If the reading is 0.08mm, the zero error is referred to as +0.08mm. If positive, the error is subtracted from the mean reading the instrument reads. Thus if the instrument reads 4.39 cm and the error is +0.05, the actual length will be 4.39-0.05=4.34 cm. If negative, the error is added to the mean reading the instrument reads. Thus if the instrument reads 4.39 cm and as above the error is -0.05 cm, the actual length will be 4.39+0.05=4.44 cm. (Considering that, the quantity is called zero correction which should always be added algebraically to the observed reading to the correct value.)
Zero Error (Z.E) = + or- n* Least Count(L.C).
See also[edit]
Micrometer
Nonius – device invented by Pedro Nunes
Transversal (instrument making) – technique in use prior to vernier scales
References[edit]
Jump up ^ Ronan, Colin A.; Needham, Joseph (24 June 1994). The Shorter Science and Civilisation in China: 4. Cambridge University Press. p. 36. ISBN 978-0-521-32995-8. adjustable outside caliper gauge... self-dated at AD 9 . An abridged version.
Jump up ^ "Bronze Caliper of the Wang Mang Regime". Archived from the original on 31 August 2014. Retrieved 26 November 2013.
Jump up ^ Barrow called the device a Vernier scale. See: John Barrow, Navigatio britannica: or a complete system of navigation … (London, England: W. and J. Mount and T. Page, 1750), pp. 140–142, especially page 142.
Jump up ^ Daumas, Maurice, Scientific Instruments of the Seventeenth and Eighteenth Centuries and Their Makers, Portman Books, London 1989 ISBN 978-0-7134-0727-3
Jump up ^ Lalande, Jérôme (1746), Astronomie, vol. 2 (Paris, France: Desaint & Saillant), pages 859-860.
Jump up ^ Davis, Raymond, Foote, Francis, Kelly, Joe, Surveying, Theory and Practice, McGraw-Hill Book Company, 1966 LC 64-66263
Jump up ^ Vernier acuity definition at the Online Medical Dictionary
Jump up ^ Kwan, A. (2011). "Vernier scales and other early devices for precise measurement". American Journal of Physics. 79 (4): 368. doi:10.1119/1.3533717.
External links[edit]
Use of vernier scale in mm and cm - simulator
Use of vernier scale in inch - simulator of measurement and interpretation
How to read a vernier caliper
Calculation of Least Count in Metric & Imperial Scales of Vernier Caliper
Classroom activity on the Vernier scale at Wikiversity
Authority control
NDL: 00568601
[hide]
v
t
e
Measuring and alignment tools
architect's scale
beam compass
calipers
chalk box
combination square
compass
coordinate-measuring machine
Diagonal scale
drafting machine
engineer's scale
flat spline
French curve
Gunter's chain
jig
laser level
laser line level
laser measuring tool
lesbian rule
measuring rod
micrometer
plumb-bob
protractor
Radius gauge
ruler
set square
sliding T bevel
speed square
spirit level
steel square
stencil
straightedge
T-square
tape measure
theodolite
Thread pitch gauge
try square
Vernier scale
vise
weighing scale
winding sticks
yardstick
Types of tools
Cleaning
Cutting and abrasive
Forestry
Garden
Hand
Machine and metalworking
Measuring and alignment
Power
Retrieved from "https://en.wikipedia.org/w/index.php?title=Vernier_scale&oldid=835811506"
Categories:
Measurement
Scales
Chinese inventions
Navigation menu
Personal tools
Not logged in
Talk
Contributions
Create account
Log in
Namespaces
Article
Talk
Variants
Views
Read
Edit
View history
More
Search
Navigation
Main page
Contents
Featured content
Current events
Random article
Donate to Wikipedia
Wikipedia store
Interaction
Help
About Wikipedia
Community portal
Recent changes
Contact page
Tools
What links here
Related changes
Upload file
Special pages
Permanent link
Page information
Wikidata item
Cite this page
Print/export
Create a book
Download as PDF
Printable version
In other projects
Wikimedia Commons
Languages
Afrikaans
العربية
Български
Bosanski
Català
Čeština
Dansk
Deutsch
Eesti
Español
Esperanto
Français
हिन्दी
Hrvatski
Ido
Bahasa Indonesia
Italiano
Kaszëbsczi
Қазақша
Latviešu
Lietuvių
Magyar
മലയാളം
Bahasa Melayu
Nederlands
日本語
Norsk
Polski
Русский
संस्कृतम्
සිංහල
Simple English
Slovenčina
Српски / srpski
Srpskohrvatski / српскохрватски
Suomi
Svenska
தமிழ்
తెలుగు
ไทย
Українська
中文
33 more
Edit links
This page was last edited on 10 April 2018, at 22:26.
Text is available under the Creative Commons Attribution-ShareAlike License; additional terms may apply. By using this site, you agree to the Terms of Use and Privacy Policy. Wikipedia® is a registered trademark of the Wikimedia Foundation, Inc., a non-profit organization.
Privacy policy
About Wikipedia
Disclaimers
Contact Wikipedia
Developers
Cookie statement
Mobile view
Enable previews

values: mw-page-base noprint

values: mw-head-base noprint

values: firstHeading firstHeading en

values: siteSub noprint

values: jump-to-nav mw-jump

value: b

values: /wiki/Jeweler mw-redirect Jeweler

values: /wiki/Caliper mw-redirect Caliper

values: Edit section: Mechanism of a vernier scale

attrs: alt src width height data-file-width data-file-height

values:
nofollow
external text
//doi.org/10.1119%2F1.3533717

values: -moz-column-width: 30em; -webkit-column-width: 30em; column-width: 30em; list-style-type: decimal;

values: Template talk:Measuring and alignment tools

values:
/wiki/Architect%27s_scale
mw-redirect
Architect's scale

values:
/wiki/Engineer%27s_scale
mw-redirect
Engineer's scale

values: nowraplinks collapsible autocollapse navbox-inner

values:
catlinks mw-normal-catlinks
catlinks mw-normal-catlinks
interface

values: Discussion about edits from this IP address [ctrl-alt-n]

values: A list of edits made from this IP address [ctrl-alt-y]

values: You are encouraged to create an account and log in; however, it is not mandatory

values: You're encouraged to log in; however, it's not mandatory. [ctrl-alt-o]

values:
/wiki/Vernier_scale
View the content page [ctrl-alt-c]
c

values: Discussion about the content page [ctrl-alt-t]

values: p-namespaces navigation vectorTabs p-namespaces-label

values:
p-variants
navigation
vectorMenu emptyPortlet
p-variants-label

values: Past revisions of this page [ctrl-alt-h]

values: p-views navigation vectorTabs p-views-label

values:
p-cactions
navigation
vectorMenu emptyPortlet
p-cactions-label

values:
mw-wiki-logo
/wiki/Main_Page
Visit the main page

values:
/wiki/Main_Page
Visit the main page [ctrl-alt-z]
z

values: Featured content – the best of Wikipedia

values: Find background information on current events

values:
/wiki/Special:Random
Load a random article [ctrl-alt-x]
x

values: portal navigation p-navigation p-navigation-label

values: Guidance on how to use and edit Wikipedia

values: About the project, what you can do, where to find things

values: A list of recent changes in the wiki [ctrl-alt-r]

values: portal navigation p-interaction p-interaction-label

values: portal navigation p-wikibase-otherprojects p-wikibase-otherprojects-label

values:
interlanguage-link interwiki-af
display: none;

values: ورنية (مقياس) – Arabic

values:
interlanguage-link interwiki-bg
display: none;

values:
interlanguage-link interwiki-bs
display: none;

values:
interlanguage-link interwiki-ca
display: none;

values:
https://cs.wikipedia.org/wiki/Vernier
Vernier – Czech
cs
cs
interlanguage-link-target

values:
interlanguage-link interwiki-cs
display: none;

values:
https://da.wikipedia.org/wiki/Nonius
Nonius – Danish
da
da
interlanguage-link-target

values:
interlanguage-link interwiki-da
display: none;

values:
https://de.wikipedia.org/wiki/Nonius
Nonius – German
de
de
interlanguage-link-target

values:
https://et.wikipedia.org/wiki/Noonius
Noonius – Estonian
et
et
interlanguage-link-target

values:
interlanguage-link interwiki-et
display: none;

values:
https://es.wikipedia.org/wiki/Nonio
Nonio – Spanish
es
es
interlanguage-link-target

values:
https://eo.wikipedia.org/wiki/Verniero
Verniero – Esperanto
eo
eo
interlanguage-link-target

values:
interlanguage-link interwiki-eo
display: none;

values: वर्नियर मापनी – Hindi

values:
https://hr.wikipedia.org/wiki/Nonius
Nonius – Croatian
hr
hr
interlanguage-link-target

values:
interlanguage-link interwiki-hr
display: none;

values:
https://io.wikipedia.org/wiki/Verniero
Verniero – Ido
io
io
interlanguage-link-target

values:
interlanguage-link interwiki-io
display: none;

values:
interlanguage-link interwiki-it
display: none;

values:
https://csb.wikipedia.org/wiki/Nonius
Nonius – Kashubian
csb
csb
interlanguage-link-target

values:
interlanguage-link interwiki-csb
display: none;

values: Нониус – Kazakh

values:
interlanguage-link interwiki-kk
display: none;

values:
https://lv.wikipedia.org/wiki/Nonijs
Nonijs – Latvian
lv
lv
interlanguage-link-target

values:
interlanguage-link interwiki-lv
display: none;

values:
interlanguage-link interwiki-lt
display: none;

values:
interlanguage-link interwiki-hu
display: none;

values: വെർണിയർ കാലിപ്പെർ – Malayalam

values:
interlanguage-link interwiki-ml
display: none;

values:
https://nl.wikipedia.org/wiki/Nonius
Nonius – Dutch
nl
nl
interlanguage-link-target

values:
interlanguage-link interwiki-nl
display: none;

values:
interlanguage-link interwiki-ja
display: none;

values:
https://no.wikipedia.org/wiki/Nonius
Nonius – Norwegian
no
no
interlanguage-link-target

values:
interlanguage-link interwiki-no
display: none;

values:
https://pl.wikipedia.org/wiki/Noniusz
Noniusz – Polish
pl
pl
interlanguage-link-target

values:
interlanguage-link interwiki-pl
display: none;

values: Нониус – Russian

values: वर्नियरमापकम् – Sanskrit

values:
interlanguage-link interwiki-sa
display: none;

values: වර්නියර් කැලිපරය – Sinhala

values:
interlanguage-link interwiki-si
display: none;

values:
interlanguage-link interwiki-simple
display: none;

values:
interlanguage-link interwiki-sk
display: none;

values: Помично кљунасто мерило – Serbian

values:
interlanguage-link interwiki-sr
display: none;

values:
interlanguage-link interwiki-sh
display: none;

values:
https://fi.wikipedia.org/wiki/Noonio
Noonio – Finnish
fi
fi
interlanguage-link-target

values:
interlanguage-link interwiki-fi
display: none;

values:
interlanguage-link interwiki-sv
display: none;

values: வேணியர் இடுக்குமானி – Tamil

values:
interlanguage-link interwiki-ta
display: none;

values: వెర్నియర్ కాలిపర్స్ – Telugu

values:
interlanguage-link interwiki-te
display: none;

values: มาตรแวร์นีเย – Thai

values:
interlanguage-link interwiki-th
display: none;

values:
interlanguage-link interwiki-uk
display: none;

values: All languages (initial selection from common choices by you and others)

values: /static/images/wikimedia-button-1.5x.png 1.5x, /static/images/wikimedia-button-2x.png 2x

values: /static/images/poweredby_mediawiki_132x47.png 1.5x, /static/images/poweredby_mediawiki_176x62.png 2x

values:
suggestions
display: none; font-size: 13px;

values: mediawiki ltr sitedir-ltr mw-hide-empty-elt ns-0 ns-subject page-Vernier_scale rootpage-Vernier_scale skin-vector action-view